Electron beam computed tomography

Short description: Medical diagnostic technology

Electron beam computed tomography

Patent illustration showing a cutaway view of an electron beam computerized tomography system. Components are 22. electron gun, 23. electron beam, 24. focus coil, 27. beam bending coil, 28-31. target rings, 14. detector array, 11. scan tube. The electron beam creates x-rays at the target rings, which radiates through the patient to the detector on the opposite end of the scan tube.

MeSH| D014057
OPS-301 code| 3-26

[edit on Wikidata]

Electron beam computed tomography (EBCT) is a specific form of computed tomography (CT) in which the X-ray tube is not mechanically spun in order to rotate the source of X-ray photons. This different design was explicitly developed to better image heart structures that never stop moving, performing a complete cycle of movement with each heartbeat.

As in conventional CT technology, the X-ray source-point moves along a circle in space around an object to be imaged. In EBCT, the X-ray tube itself is large and stationary, and partially surrounds the imaging circle. Rather than moving the tube itself, electron-beam focal point (and hence the X-ray source point) is swept electronically along a tungsten anode in the tube, tracing a large circular arc on its inner surface. This motion can be very fast.

## Contents

* 1 Advantages and disadvantages
* 2 Design specifics
* 3 Future
* 4 References

## Advantages and disadvantages

The principal application advantage of EBT machines, and the reason for the invention, is that because the X-ray source-point is swept electronically, not mechanically, it can be swept with far greater speed.

The major medical application for which this design technology was invented in the 1980s was for imaging the human heart, specifically to detect coronary calcium.[1] The heart never stops moving, and some important structures, such as arteries, move several times their diameter during each heartbeat. Rapid imaging is therefore important to prevent blurring of moving structures during the scan. EBT detection of calcium deposits is accurate, fast and involves lower exposure to ionising radiation than conventional CT.[2][3] Patients are exposed to radiation for a shorter period as it is faster in creating multiple images of the heart. The most advanced current commercial designs can perform image sweeps in as little as 0.025 seconds. By comparison, the fastest mechanically swept X-ray tube designs require about 0.25 seconds to perform an image sweep.[4] For reference, current coronary artery angiography imaging is usually performed at 30 frames/second or 0.033 seconds/frame; EBT is far closer to this than mechanically swept CT machines.

Signal-to-noise ratio and spatial resolution are also worse compared to conventional CT.[3]

## Design specifics

As in standard X-ray tubes, part of the electron current energy when hitting the tungsten target is converted into photons. However, instead of spinning a small target anode in order to dissipate waste heat, the electron current focus spot is swept along a large stationary target anode.[5]

The electron current sweep is aimed using wound copper coil magnetic deflection yokes, as in a cathode ray tube (CRT). However, the entire structure of the cathode, deflection yokes, anode and overall vacuum tube size is much larger, therefore made out of steel, not glass, with the main central open midsection of the vacuum tube hollow, leaving room for the scan table and object or person to lie while the scan is performed.

## Future

Whether the inherent sweep-speed advantage will maintain commercial viability of the EBT design remains unclear at this time. As of 2002, one major company owns and offers models in both competing designs, with engineering cross-pollination of techniques between the product design teams. As of 2005, it increasingly appears that the spiral CT designs, especially those with 64 detector rows, 3×360°/sec rotation speeds, and designed for cardiac imaging, are largely replacing the EBT design from a commercial and medical perspective. However, EBT still offers sweep speeds of effectively 50×360°/sec rotation speeds and lower radiation exposure. The latest version of the EBT eSpeed offers a 33 ms sweep time.

This technology continues to represent the fastest commercial CT temporal resolution.

Since 2008, a single development company has taken the lead on continued development, support and sales of EBT imaging products. The EBT scanner continues to stay in use globally due to the high accuracy, superior repeatability and ultra low dose abilities over high dose mechanical scanners.

## References

1. ↑ Mittal, Tarun K.; Rubens, Michael B. (2006). "Computed Tomography Techniques and Principles. Part a. Electron Beam Computed Tomography". in Anagnostopoulos, Constantinos D.; Bax, Jeroen J.; Nihoyannopoulos, Petros et al.. Noninvasive Imaging of Myocardial Ischemia. New York: Springer-Verlag. pp. 93. doi:10.1007/1-84628-156-3_6. ISBN 978-1-84628-027-6.
2. ↑ Raggi, Paolo (January 2001). "Imaging of cardiovascular calcifications with electron beam tomography in hemodialysis patients". American Journal of Kidney Diseases 37 (1): S62–S65. doi:10.1053/ajkd.2001.20745.
3. ↑ 3.0 3.1 Peebles, C R (1 June 2003). "Non-invasive coronary imaging: computed tomography or magnetic resonance imaging?". Heart 89 (6): 591–594. doi:10.1136/heart.89.6.591. PMID 12748207.
4. ↑ "SOMATOM Force" (in en). https://www.healthcare.siemens.co.uk/computed-tomography/dual-source-ct/somatom-force/technical-specifications.
5. ↑ Hill, David G. (2005). "Electron Beam CT of the Heart". in Schoepf, U. Joseph. CT of the Heart. Totowa, N.J.: Humana Press. pp. 15–21. doi:10.1385/1-59259-818-8:015. ISBN 978-1-58829-303-9.

* v
* t
* e

Medical imaging (ICD-9-CM V3 87–88, ICD-10-PCS B, CPT 70010–79999)

X-ray/
Radiography|

| 2D|

| Medical:|

* Pneumoencephalography
* Dental radiography
* Sialography
* Myelography
* CXR
* Bronchography
* AXR
* KUB
* DXA/DXR
* Upper gastrointestinal series/Small-bowel follow-through/Lower gastrointestinal series
* Cholangiography/Cholecystography
* Mammography
* Pyelogram
* Cystography
* Arthrogram
* Hysterosalpingography
* Skeletal survey
* Angiography
* Angiocardiography
* Aortography
* Venography
* Lymphogram
* Orbital radiography

|

Industrial:|

* Radiographic testing

CT scan|

| Techniques:|

* General operation of CT
* Quantitative CT
* High-resolution CT
* X-ray microtomography
* Electron beam computed tomography
* Cone beam computed tomography

|

Targets|

* Coronary
* Calcium scan
* CT angiography
* Abdominal and pelvic CT
* Virtual colonoscopy
* CT angiography
* Coronary CT
* Pulmonary CT
* Head CT
* Thyroid CT
* Whole body imaging
* Full-body CT scan

Other|

* Fluoroscopy
* Dental panoramic radiography
* X-ray motion analysis

MRI|

* MRI of the brain
* MR neurography
* Cardiac MRI/Cardiac MRI perfusion
* MR angiography
* MR cholangiopancreatography
* Breast MRI
* Functional MRI
* Sequences
* Diffusion MRI
* Perfusion MRI
* Tractography
* Synthetic MRI

Ultrasound|

* Echocardiography
* Doppler ultrasonography
* Doppler echocardiography
* TTE
* TEE
* Transcranial Doppler
* Intravascular
* Gynecologic
* Obstetric
* Echoencephalography
* Abdominal ultrasonography
* Transrectal
* Breast ultrasound
* Transscrotal ultrasound
* Carotid ultrasonography
* Contrast-enhanced
* 3D ultrasound
* Endoscopic ultrasound
* Emergency ultrasound
* FAST
* Pre-hospital ultrasound
* Duplex

Radionuclide|

| 2D / scintigraphy|

* Cholescintigraphy
* Scintimammography
* Ventilation/perfusion scan
* Radionuclide ventriculography
* Radionuclide angiography
* Radioisotope renography
* Sestamibi parathyroid scintigraphy
* Radioactive iodine uptake test
* Bone scintigraphy
* Immunoscintigraphy
* Dacryoscintigraphy
* DMSA scan
* Gastric emptying scan

Full body:|

* Octreotide scan
* Gallium 67 scan
* Indium-111 WBC scan

3D / ECT|

| SPECT (gamma ray):|

* Myocardial perfusion imaging

|

PET (positron):|

* Brain PET
* Cardiac PET
* PET mammography
* PET-CT
* PET-MRI

Optical/Laser|

* Optical tomography
* Optical coherence tomography
* Confocal microscopy
* Endomicroscopy
* Orthogonal polarization spectral imaging

Thermography|

* non-contact thermography
* contact thermography
* dynamic angiothermography

Target conditions|

* Acute stroke
* Pregnancy

* Category

0.00

(0 votes)

Original source: https://en.wikipedia.org/wiki/Electron beam computed tomography. Read more

|

Retrieved from "https://handwiki.org/wiki/index.php?title=Astronomy:Electron_beam_computed_tomography&oldid=679515"
*[v]: View this template
*[t]: Discuss this template
*[e]: Edit this template