Carbon dioxide sensor

Instrument for the measurement of carbon dioxide gas

Not to be confused with Carbon monoxide detector.

A carbon dioxide sensor or CO2 sensor is an instrument for the measurement of carbon dioxide gas. The most common principles for CO2 sensors are infrared gas sensors (NDIR) and chemical gas sensors. Measuring carbon dioxide is important in monitoring indoor air quality,[1] the function of the lungs in the form of a capnograph device, and many industrial processes.

## Nondispersive infrared (NDIR) CO2 sensors

[edit]

Main article: Nondispersive infrared sensor

See also: Capnography

CO2 concentration meter using a nondispersive infrared sensor

NDIR sensors are spectroscopic sensors to detect CO2 in a gaseous environment by its characteristic absorption. The key components are an infrared source, a light tube, an interference (wavelength) filter, and an infrared detector. The gas is pumped or diffuses into the light tube, and the electronics measure the absorption of the characteristic wavelength of light. NDIR sensors are most often used for measuring carbon dioxide.[2] The best of these have sensitivities of 20–50 PPM.[2] Typical NDIR sensors cost in the (US) $100 to $1000 range.

NDIR CO2 sensors are also used for dissolved CO2 for applications such as beverage carbonation, pharmaceutical fermentation and CO2 sequestration applications. In this case they are mated to an ATR (attenuated total reflection) optic and measure the gas in situ. New developments include using microelectromechanical systems (MEMS) IR sources to bring down the costs of this sensor and to create smaller devices (for example for use in air conditioning applications).[3]

Another method (Henry's Law) also can be used to measure the amount of dissolved CO2 in a liquid, if the amount of foreign gases is insignificant.[further explanation needed]

## Photoacoustic sensors

[edit]

CO2 can be measured using photoacoustic spectroscopy. Concentration of CO2 can be measured by subjecting a sample to pulses of electromagnetic energy (such as from a distributed feedback laser[4]) that is tuned specifically to the absorption wavelength of CO2. With each pulse of energy, the CO2 molecules within the sample will absorb and generate pressure waves via the photoacoustic effect. These pressure waves are then detected with an acoustic detector and converted to a usable CO2 reading through a computer or microprocessor.[5]

## Chemical CO2 sensors

[edit]

Chemical CO2 gas sensors with sensitive layers based on polymer- or heteropolysiloxane have the principal advantage of very low energy consumption, and that they can be reduced in size to fit into microelectronic-based systems. On the downside, short and long term drift effects, as well as a rather low overall lifetime, are major obstacles when compared with the NDIR measurement principle.[6] Most CO2 sensors are fully calibrated prior to shipping from the factory. Over time, the zero point of the sensor needs to be calibrated to maintain the long term stability of the sensor.[7]

## Estimated CO2 sensor

[edit]

For indoor environments such as offices or gyms where the principal source of CO2 is human respiration, rescaling some easier-to-measure quantities such as volatile organic compound (VOC) and hydrogen gas (H2) concentrations provides a good-enough estimator of the real CO2 concentration for ventilation and occupancy purposes.[citation needed] Furthermore, inasmuch as ventilation is a factor in the spread of respiratory viruses,[8] CO2 levels are a rough metric for COVID-19 risk; the worse the ventilation, the better for viruses and vice versa.[9] Sensors for these substances can be made using cheap (~$20) Microelectromechanical systems (MEMS) metal oxide semiconductor (MOS) technology. The reading they generate is called estimated CO2 (eCO2)[10] or CO2 equivalent (CO2eq).[11] Although the readings tend to be good enough in the long run, introducing non-respiration sources of VOC or CO2, such as peeling fruits or using perfume, will undermine their reliability. H2-based sensors are less susceptible as they are more specific to human breathing, although the very health conditions the hydrogen breath test is set to diagnose will also disrupt them.[11]

## Applications

[edit]

* Examples:
* Modified atmospheres
* Indoor air quality
* Stowaway detection
* Cellar and gas stores
* Marine vessels
* Greenhouses
* Landfill gas
* Confined spaces
* Aerospace
* Healthcare
* Horticulture
* Transportation
* Cryogenics
* Ventilation management
* Mining
* Rebreathers (SCUBA)
* Decaffeination
* For indoor human occupancy counting[12][13]
* For HVAC applications, CO2 sensors can be used to monitor the quality of air and the tailored need for fresh air, respectively. Measuring CO2 levels indirectly determines how many people are in a room, and ventilation can be adjusted accordingly. See demand controlled ventilation (DCV).[14]

## See also

[edit]

* Exhaust gas analyzer
* Oxygen sensor
* Gas detector
* Colorimetric capnography
* Carbon dioxide recorder

## References

[edit]

1. ^ Kampezidou, S. I.; Tikayat Ray, A.; Duncan, S.; Balchanos, M.G.; Mavris, D.N. (2021-01-07). "Real-time occupancy detection with physics-informed pattern-recognition machines based on limited CO2 and temperature sensors". Energy and Buildings. 242: 110863. Bibcode:2021EneBu.24210863K. doi:10.1016/j.enbuild.2021.110863. ISSN 0378-7788. S2CID 233831299.
2. ^ a b Lang, T.; Wiemhöfer, H.D.; Göpel, W. (1996). "Carbonate based CO2 sensors with high performance". Sensors and Actuators B: Chemical. 34 (1–3): 383–7\. doi:10.1016/S0925-4005(96)01846-1.1–3&rft.pages=383-7&rft.date=1996&rft_id=info:doi/10.1016/S0925-4005(96)01846-1&rft.aulast=Lang&rft.aufirst=T.&rft.au=Wiemhöfer,+H.D.&rft.au=Göpel,+W.&rfr_id=info:sid/en.wikipedia.org:Carbon+dioxide+sensor" class="Z3988">
3. ^ Vincent, T.A.; Gardner, J.W. (November 2016). "A low cost MEMS based NDIR system for the monitoring of carbon dioxide in breath analysis at ppm levels". Sensors and Actuators B: Chemical. 236: 954–964\. Bibcode:2016SeAcB.236..954V. doi:10.1016/j.snb.2016.04.016.
4. ^ Zakaria, Ryadh (March 2010). "3.5 Photoacoustic Spectroscopy (PAS)" (PDF). NDIR Instrumentation Design for CO2 Gas Sensing (PhD). Cranfield University. pp. 35–36\. hdl:1826/6784.
5. ^ AG, Infineon Technologies. "CO2 Sensors - Infineon Technologies". www.infineon.com. Retrieved 2020-11-10.2+Sensors+-+Infineon+Technologies&rft.aulast=AG&rft.aufirst=Infineon+Technologies&rft_id=https://www.infineon.com/cms/en/product/sensor/co2-sensors/&rfr_id=info:sid/en.wikipedia.org:Carbon+dioxide+sensor" class="Z3988">
6. ^ Zhou, R.; Vaihinger, S.; Geckeler, K.E.; Göpel, W. (1994). "Reliable CO2 sensors with silicon-based polymers on quartz microbalance transducers". Sensors and Actuators B: Chemical. 19 (1–3): 415–420\. doi:10.1016/0925-4005(93)01018-Y.1–3&rft.pages=415-420&rft.date=1994&rft_id=info:doi/10.1016/0925-4005(93)01018-Y&rft.aulast=Zhou&rft.aufirst=R.&rft.au=Vaihinger,+S.&rft.au=Geckeler,+K.E.&rft.au=Göpel,+W.&rfr_id=info:sid/en.wikipedia.org:Carbon+dioxide+sensor" class="Z3988">
7. ^ "CO2 Auto-Calibration Guide" (PDF). Archived from the original (PDF) on 2014-08-19. Retrieved 2014-08-19.
8. ^ Moriyama, Miyu; Hugentobler, Walter J.; Iwasaki, Akiko (29 September 2020). "Seasonality of Respiratory Viral Infections". Annual Review of Virology. 7 (1): 83–101\. doi:10.1146/annurev-virology-012420-022445. PMID 32196426. S2CID 214601321.
9. ^ Peng, Zhe; Jimenez, Jose L. (11 May 2021). "Exhaled CO 2 as a COVID-19 Infection Risk Proxy for Different Indoor Environments and Activities". Environmental Science & Technology Letters. 8 (5): 392–397\. Bibcode:2021EnSTL...8..392P. doi:10.1021/acs.estlett.1c00183. PMC 8043197. PMID 37566374.
10. ^ Rüffer, D; Hoehne, F; Bühler, J (31 March 2018). "New Digital Metal-Oxide (MOx) Sensor Platform". Sensors (Basel, Switzerland). 18 (4): 1052. Bibcode:2018Senso..18.1052.. doi:10.3390/s18041052. PMC 5948493. PMID 29614746.
11. ^ a b Herberger S, Herold M, Ulmer H (2009). "MOS gas sensor technology for demand controlled ventilation" (PDF). Proceedings of the 4th International Symposium on Building and Ductwork Air Tightness and 30th AIVC Conference on Trends in High Performance Buildings and the Role of Ventilation. Berlin.
12. ^ Arief-Ang, I.B.; Hamilton, M.; Salim, F. (2018-06-01). "RUP: Large Room Utilisation Prediction with carbon dioxide sensor". Pervasive and Mobile Computing. 46: 49–72\. doi:10.1016/j.pmcj.2018.03.001. ISSN 1873-1589. S2CID 13670861.
13. ^ Arief-Ang, I.B.; Salim, F.D.; Hamilton, M. (2018-04-14). "SD-HOC: Seasonal Decomposition Algorithm for Mining Lagged Time Series". Data Mining [SD-HOC: Seasonal Decomposition Algorithm for Mining Lagged Time Series]. Communications in Computer and Information Science. Vol. 845\. Springer. pp. 125–143\. doi:10.1007/978-981-13-0292-3_8. ISBN 978-981-13-0291-6.
14. ^ "Demand Control Ventilation Benefits for Your Building" (PDF). KMC Controls. 2013. Archived from the original (PDF) on 2014-06-27.

* v
* t
* e

Heating, ventilation, and air conditioning

Fundamental
concepts|

* Air changes per hour (ACH)
* Bake-out
* Building envelope
* Convection
* Dilution
* Domestic energy consumption
* Enthalpy
* Fluid dynamics
* Gas compressor
* Heat pump and refrigeration cycle
* Heat transfer
* Humidity
* Infiltration
* Latent heat
* Noise control
* Outgassing
* Particulates
* Psychrometrics
* Sensible heat
* Stack effect
* Thermal comfort
* Thermal destratification
* Thermal mass
* Thermodynamics
* Vapour pressure of water

Technology|

* Absorption-compression heat pump
* Absorption refrigerator
* Air barrier
* Air conditioning
* Antifreeze
* Automobile air conditioning
* Autonomous building
* Building insulation materials
* Central heating
* Central solar heating
* Chilled beam
* Chilled water
* Constant air volume (CAV)
* Coolant
* Cross ventilation
* Dedicated outdoor air system (DOAS)
* Deep water source cooling
* Demand controlled ventilation (DCV)
* Displacement ventilation
* District cooling
* District heating
* Electric heating
* Energy recovery ventilation (ERV)
* Firestop
* Forced-air
* Forced-air gas
* Free cooling
* Heat recovery ventilation (HRV)
* Hybrid heat
* Hydronics
* Ice storage air conditioning
* Kitchen ventilation
* Mixed-mode ventilation
* Microgeneration
* Passive cooling
* Passive daytime radiative cooling
* Passive house
* Passive ventilation
* Radiant heating and cooling
* Radiant cooling
* Radiant heating
* Radon mitigation
* Refrigeration
* Renewable heat
* Room air distribution
* Solar air heat
* Solar combisystem
* Solar cooling
* Solar heating
* Thermal insulation
* Thermosiphon
* Underfloor air distribution
* Underfloor heating
* Vapor barrier
* Vapor-compression refrigeration (VCRS)
* Variable air volume (VAV)
* Variable refrigerant flow (VRF)
* Ventilation
* Water heat recycling

Components|

* Air conditioner inverter
* Air door
* Air filter
* Air handler
* Air ionizer
* Air-mixing plenum
* Air purifier
* Air source heat pump
* Attic fan
* Automatic balancing valve
* Back boiler
* Barrier pipe
* Blast damper
* Boiler
* Centrifugal fan
* Ceramic heater
* Chiller
* Condensate pump
* Condenser
* Condensing boiler
* Convection heater
* Compressor
* Cooling tower
* Damper
* Dehumidifier
* Duct
* Economizer
* Electrostatic precipitator
* Evaporative cooler
* Evaporator
* Exhaust hood
* Expansion tank
* Fan
* Fan coil unit
* Fan filter unit
* Fan heater
* Fire damper
* Fireplace
* Fireplace insert
* Freeze stat
* Flue
* Freon
* Fume hood
* Furnace
* Gas compressor
* Gas heater
* Gasoline heater
* Grease duct
* Grille
* Ground-coupled heat exchanger
* Ground source heat pump
* Heat exchanger
* Heat pipe
* Heat pump
* Heating film
* Heating system
* HEPA
* High efficiency glandless circulating pump
* High-pressure cut-off switch
* Humidifier
* Infrared heater
* Inverter compressor
* Kerosene heater
* Louver
* Mechanical room
* Oil heater
* Packaged terminal air conditioner
* Plenum space
* Pressurisation ductwork
* Process duct work
* Radiator
* Radiator reflector
* Recuperator
* Refrigerant
* Register
* Reversing valve
* Run-around coil
* Sail switch
* Scroll compressor
* Solar chimney
* Solar-assisted heat pump
* Space heater
* Smoke canopy
* Smoke damper
* Smoke exhaust ductwork
* Thermal expansion valve
* Thermal wheel
* Thermostatic radiator valve
* Trickle vent
* Trombe wall
* TurboSwing
* Turning vanes
* Ultra-low particulate air (ULPA)
* Whole-house fan
* Windcatcher
* Wood-burning stove
* Zone valve

Measurement
and control|

* Air flow meter
* Aquastat
* BACnet
* Blower door
* Building automation
* Carbon dioxide sensor
* Clean air delivery rate (CADR)
* Control valve
* Gas detector
* Home energy monitor
* Humidistat
* HVAC control system
* Infrared thermometer
* Intelligent buildings
* LonWorks
* Minimum efficiency reporting value (MERV)
* Normal temperature and pressure (NTP)
* OpenTherm
* Programmable communicating thermostat
* Programmable thermostat
* Psychrometrics
* Room temperature
* Smart thermostat
* Standard temperature and pressure (STP)
* Thermographic camera
* Thermostat
* Thermostatic radiator valve

Professions,
trades,
and services|

* Architectural acoustics
* Architectural engineering
* Architectural technologist
* Building services engineering
* Building information modeling (BIM)
* Deep energy retrofit
* Duct cleaning
* Duct leakage testing
* Environmental engineering
* Hydronic balancing
* Kitchen exhaust cleaning
* Mechanical engineering
* Mechanical, electrical, and plumbing
* Mold growth, assessment, and remediation
* Refrigerant reclamation
* Testing, adjusting, balancing

Industry
organizations|

* AHRI
* AMCA
* ASHRAE
* ASTM International
* BRE
* BSRIA
* CIBSE
* Institute of Refrigeration
* IIR
* LEED
* SMACNA
* UMC

Health and safety|

* Indoor air quality (IAQ)
* Passive smoking
* Sick building syndrome (SBS)
* Volatile organic compound (VOC)

See also|

* ASHRAE Handbook
* Building science
* Fireproofing
* Glossary of HVAC terms
* Warm Spaces
* World Refrigeration Day
* Template:Fire protection
* Template:Home automation
* Template:Solar energy

*[v]: View this template
*[t]: Discuss this template
*[e]: Edit this template