Dual-polarization interferometry

Analytical technique that probes molecular layers adsorbed to the surface of a waveguide

Dual-polarization interferometry (DPI) is an analytical technique that probes molecular layers adsorbed to the surface of a waveguide using the evanescent wave of a laser beam. It is used to measure the conformational change in proteins, or other biomolecules, as they function (referred to as the conformation activity relationship).

## Instrumentation

[edit]

DPI[1] focuses laser light into two waveguides. One of these functions as the "sensing" waveguide having an exposed surface while the second one functions to maintain a reference beam. A two-dimensional interference pattern is formed in the far field by combining the light passing through the two waveguides. The DPI technique rotates the polarization of the laser, to alternately excite two polarization modes of the waveguides. Measurement of the interferogram for both polarizations allows both the refractive index and the thickness of the adsorbed layer to be calculated. The polarization can be switched rapidly, allowing real-time measurements of chemical reactions taking place on a chip surface in a flow-through system. These measurements can be used to infer conformational information about the molecular interactions taking place, as the molecule size (from the layer thickness) and the fold density (from the RI) change. DPI is typically used to characterize biochemical interactions by quantifying any conformational change at the same time as measuring reaction rates, affinities and thermodynamics.[citation needed]

The technique is quantitative and real-time (10 Hz) with a dimensional resolution of 0.01 nm.[2]

Extensions of dual-polarization interferometry also exist, namely multiple pathlength dual-polarization interferometry (MPL-DPI)[3][4][5] and absorption enhanced DPI. In MPL-DPI quantification of both layer thickness and refractive index (density) and therefore mass per unit area can be made for in situ and ex-situ coated films, where normal DPI can only calculate film properties if the interferogram is constantly monitored. Absorption enhanced DPI[6] (AE-DPI) is used to separate the mass of different molecules on the surface, exploiting the absorption of one of the molecular species compared to the other species on the surface.

## Applications

[edit]

A novel application for dual-polarization interferometry emerged in 2008, where the intensity of light passing through the waveguide is extinguished in the presence of crystal growth. This has allowed the very earliest stages in protein crystal nucleation to be monitored.[7] Later versions of dual-polarization interferometers also have the capability to quantify the order and disruption in birefringent thin films.[8] This has been used, for example, to study the formation of lipid bilayers and their interaction with membrane proteins.[9][10]

## References

[edit]

1. ^ Cross, G; Reeves, AA; Brand, S; Popplewell, JF; Peel, LL; Swann, MJ; Freeman, NJ (2003). "A new quantitative optical biosensor for protein characterisation". Biosensors and Bioelectronics. 19 (4): 383–90\. doi:10.1016/S0956-5663(03)00203-3. PMID 14615097.
2. ^ Swann, MJ; Freeman, NJ; Cross, GH (2007). "Dual Polarization Interferometry: A Real-Time Optical Technique for Measuring (Bio)Molecular Orientation, Structure and Function at the Solid/Liquid Interface". In Marks, R.S.; Lowe, C.R.; Cullen, D.C.; Weetall, H.H.; Karube, I. (eds.). Handbook of Biosensors and Biochips. Vol. 1\. Wiley. Pt. 4, Ch. 33, pp. 549–568. ISBN 978-0-470-01905-4.
3. ^ Coffey, Paul D.; Swann, Marcus J.; Waigh, Thomas A.; Schedin, Fred; Lu, Jian R. (June 22, 2009). "Multiple path length dual polarization interferometry". Optics Express. 17 (13): 10959–10969\. Bibcode:2009OExpr..1710959C. doi:10.1364/OE.17.010959. PMID 19550495.
4. ^ Coffey, Paul. "Interfacial Measurements of Colloidal and Bio-colloidal Systems in Real-Time". Retrieved December 16, 2022.
5. ^ Delivopoulos, Evangelos; Ouberai, Myriam M.; Coffey, Paul D.; Swann, Marcus J.; Shakesheff, Kevin M.; Welland, Mark E. (February 2015). "Serum protein layers on parylene-C and silicon oxide: Effect on cell adhesion". Colloids and Surfaces B: Biointerfaces. 126: 169–177\. doi:10.1016/j.colsurfb.2014.12.020. PMC 4342411. PMID 25555155.
6. ^ Coffey, Paul David; Swann, Marcus Jack; Waigh, Thomas Andrew; Mu, Qingshan; Lu, Jian Ren (2013). "The structure and mass of heterogeneous thin films measured with dual polarization interferometry and ellipsometry". RSC Advances. 3 (10): 3316. Bibcode:2013RSCAd...3.3316C. doi:10.1039/C2RA22911K.
7. ^ Boudjemline, A; Clarke, DT; Freeman, NJ; Nicholson, JM; Jones, GR (2008). "Early stages of protein crystallization as revealed by emerging optical waveguide technology". Journal of Applied Crystallography. 41 (3): 523. doi:10.1107/S0021889808005098.
8. ^ Mashaghi, A; Swann, M; Popplewell, J; Textor, M; Reimhult, E (2008). "Optical Anisotropy of Supported Lipid Structures Probed by Waveguide Spectroscopy and Its Application to Study of Supported Lipid Bilayer Formation Kinetics". Analytical Chemistry. 80 (10): 3666–76\. doi:10.1021/ac800027s. PMID 18422336.
9. ^ Sanghera, N; Swann, MJ; Ronan, G; Pinheiro, TJ (2009). "Insight into early events in the aggregation of the prion protein on lipid membranes". Biochimica et Biophysica Acta (BBA) - Biomembranes. 1788 (10): 2245–51\. doi:10.1016/j.bbamem.2009.08.005. PMID 19703409.
10. ^ Lee, TH; Heng, C; Swann, MJ; Gehman, JD; Separovic, F; Aguilar, MI (2010). "Real-time quantitative analysis of lipid disordering by aurein 1.2 during membrane adsorption, destabilisation and lysis". Biochimica et Biophysica Acta (BBA) - Biomembranes. 1798 (10): 1977–86\. doi:10.1016/j.bbamem.2010.06.023. PMID 20599687.

## Further reading

[edit]

* Cross, GH; Ren, Y; Freeman, NJ (1999). "Young's fringes from vertically integrated slab waveguides: Applications to humidity sensing" (PDF). Journal of Applied Physics. 86 (11): 6483. Bibcode:1999JAP....86.6483C. doi:10.1063/1.371712. S2CID 121706760.
* Cross, G (2003). "A new quantitative optical biosensor for protein characterisation". Biosensors and Bioelectronics. 19 (4): 383–90\. doi:10.1016/S0956-5663(03)00203-3. PMID 14615097.
* Freeman, NJ; Peel, LL; Swann, MJ; Cross, GH; Reeves, A; Brand, S; Lu, JR (2004). "Real time, high resolution studies of protein adsorption and structure at the solid–liquid interface using dual polarization interferometry". Journal of Physics: Condensed Matter. 16 (26): S2493 – S2496. Bibcode:2004JPCM...16S2493F. doi:10.1088/0953-8984/16/26/023. S2CID 250737643.
* Khan, TR; Grandin, HM; Mashaghi, A; Textor, M; Reimhult, E; Reviakine, I (2008). "Lipid redistribution in phosphatidylserine-containing vesicles adsorbing on titania". Biointerphases. 3 (2): FA90. doi:10.1116/1.2912098. PMID 20408675. S2CID 28632810.

* v
* t
* e

Spectroscopy

Vibrational (IR)|

* FT-IR
* Raman
* Resonance Raman
* Rotational
* Rotational–vibrational
* Vibrational
* Vibrational circular dichroism
* Nuclear resonance vibrational spectroscopy
* Vibrational spectroscopy of linear molecules
* Thermal infrared spectroscopy

UV–Vis–NIR "Optical"|

* Ultraviolet–visible
* Fluorescence
* Cold vapour atomic
* Vibronic
* Near-infrared
* Resonance-enhanced multiphoton ionization (REMPI)
* Raman
* Coherent anti-Stokes
* Raman optical activity
* Laser-induced breakdown
* Atomic
* emission
* Glow-discharge optical
* absorption
* Cavity ring-down spectroscopy
* Saturated absorption spectroscopy

X-ray and Gamma ray|

* X-ray
* Energy-dispersive
* Emission
* Extended X-ray absorption fine structure
* Gamma
* Mössbauer
* Conversion electron

Electron|

* Photoelectron/photoemission
* X-ray
* UV
* Angle-resolved
* Two-photon
* Auger
* phenomenological
* paramagnetic
* Beta spectroscopy

Nucleon|

* Alpha
* Inelastic neutron scattering
* Neutron spin echo

Radiowave|

* NMR
* 2D
* Terahertz
* ESR/EPR
* Ferromagnetic resonance

Others|

| Data collection, processing|

* Fourier-transform spectroscopy
* Hyperspectral imaging
* Spectrophotometry
* Time-stretch
* Time-resolved spectroscopy
* Video spectroscopy

|

Measured phenomena|

* Acoustic resonance spectroscopy
* Circular dichroism spectroscopy
* Deep-level transient spectroscopy
* Dual-polarization interferometry
* Hadron spectroscopy
* Inelastic electron tunneling spectroscopy
* Scanning tunneling spectroscopy
* Photoacoustic spectroscopy
* Photothermal spectroscopy
* Pump–probe spectroscopy

Applications|

* Astronomical spectroscopy
* Force spectroscopy (a misnomer)

* Category
* Commons

* v
* t
* e

Protein structural analysis

High resolution|

* Cryogenic electron microscopy
* X-ray crystallography
* NMR
* Electron crystallography
* EPR

Medium resolution|

* Fiber diffraction
* Mass spectrometry
* SAXS

Spectroscopic|

* NMR
* Circular dichroism
* Dual-polarization interferometry
* Absorbance
* Fluorescence
* Fluorescence anisotropy

Translational diffusion|

* Analytical ultracentrifugation
* Size exclusion chromatography
* Light scattering
* NMR

Rotational diffusion|

* Fluorescence anisotropy
* Flow birefringence
* Dielectric relaxation
* NMR

Chemical|

* Hydrogen–deuterium exchange
* Site-directed mutagenesis
* Chemical modification

Thermodynamic|

* Equilibrium unfolding

Computational|

* Protein structure prediction
* Molecular docking

←Tertiary structure

Quaternary structure→
*[v]: View this template
*[t]: Discuss this template
*[e]: Edit this template