Functional near-infrared spectroscopy

Optical technique for monitoring brain activity

fNIRS with a Gowerlabs NTS system

Functional near-infrared spectroscopy (fNIRS) is an optical brain monitoring technique which uses near-infrared spectroscopy for the purpose of functional neuroimaging.[1] Using fNIRS, brain activity is measured by using near-infrared light to estimate cortical hemodynamic activity which occur in response to neural activity. Alongside EEG, fNIRS is one of the most common non-invasive neuroimaging techniques which can be used in portable contexts. The signal is often compared with the BOLD signal measured by fMRI and is capable of measuring changes both in oxy- and deoxyhemoglobin concentration,[2] but can only measure from regions near the cortical surface. fNIRS may also be referred to as Optical Topography (OT) and is sometimes referred to simply as NIRS.

## Description

[edit]

Oxygenated and deoxygenated hemoglobin

fNIRS estimates the concentration of hemoglobin from changes in absorption of near infrared light. As light moves or propagates through the head, it is alternately scattered or absorbed by the tissue through which it travels. Because hemoglobin is a significant absorber of near-infrared light, changes in absorbed light can be used to reliably measure changes in hemoglobin concentration. Different fNIRS techniques can also use the way in which light propagates to estimate blood volume and oxygenation. The technique is safe, non-invasive, and can be used with other imaging modalities.

fNIRS is a non-invasive imaging method involving the quantification of chromophore concentration resolved from the measurement of near infrared (NIR) light attenuation or temporal or phasic changes. The technique takes advantage of the optical window in which (a) skin, tissue, and bone are mostly transparent to NIR light (700–900 nm spectral interval) and (b) hemoglobin (Hb) and deoxygenated-hemoglobin (deoxy-Hb) are strong absorbers of light.

Absorption spectra for oxy-Hb and deoxy-Hb for near-infrared wavelengths

There are six different ways for infrared light to interact with the brain tissue: direct transmission, diffuse transmission, specular reflection, diffuse reflection, scattering, and absorption. fNIRS focuses primarily on absorption: differences in the absorption spectra of deoxy-Hb and oxy-Hb allow the measurement of relative changes in hemoglobin concentration through the use of light attenuation at multiple wavelengths. Two or more wavelengths are selected, with one wavelength above and one below the isosbestic point of 810 nm—at which deoxy-Hb and oxy-Hb have identical absorption coefficients. Using the modified Beer-Lambert law (mBLL), relative changes in concentration can be calculated as a function of total photon path length.[3]

Typically, the light emitter and detector are placed ipsilaterally (each emitter/detector pair on the same side) on the subject's skull so recorded measurements are due to back-scattered (reflected) light following elliptical pathways.[4] fNIRS is most sensitive to hemodynamic changes which occur nearest to the scalp[5] and these superficial artifacts are often addressed using additional light detectors located closer to the light source (short-separation detectors).[6]

### Modified Beer–Lambert law

[edit]

Changes in light intensity can be related to changes in relative concentrations of hemoglobin through the modified Beer–Lambert law (mBLL). The Beer lambert-law has to deal with concentration of hemoglobin. This technique also measures relative changes in light attenuation as well as using mBLL to quantify hemoglobin concentration changes.[7]

Basic functional near infrared spectroscopy (fNIRS) abbreviations

BFi = blood flow index

CBF = cerebral blood flow

CBV = cerebral blood volume

CMRO2= metabolic rate of oxygen

CW= continuous wave

DCS = diffuse correlation spectroscopy

FD = frequency-domain

Hb, HbR= deoxygenated hemoglobin

HbO, HbO2= oxygenated hemoglobin

HbT= total hemoglobin concentration

HGB = blood hemoglobin

SaO2= arterial saturation

SO2= hemoglobin saturation

SvO2= venous saturation

TD=time-domain

## History

[edit]

### US & UK

[edit]

In 1977, Jöbsis[8] reported that brain tissue transparency to NIR light allowed a non-invasive and continuous method of tissue oxygen saturation using transillumination. Transillumination (forward-scattering) was of limited utility in adults because of light attenuation and was quickly replaced by reflectance-mode based techniques - resulting in development of NIRS systems proceeding rapidly. Then, by 1985, the first studies on cerebral oxygenation were conducted by M. Ferrari. Later, in 1989, following work with David Delpy at University College London, Hamamatsu developed the first commercial NIRS system: NIR-1000 cerebral oxygenation monitor. NIRS methods were initially used for cerebral oximetry in the 1990s. In 1993, four publications by Chance et al. PNAS, Hoshi & Tamura J Appl Physiol, Kato et al. JCBFM, Villringer et al Neuros. Lett. demonstrated the feasibility of fNIRS in adult humans. NIRS techniques were further expanded on by the work of Randall Barbour, Britton Chance, Arno Villringer, M. Cope, D. T. Delpy, Enrico Gratton, and others. Currently, wearable fNIRS are being developed.

Hitachi ETG-4000

### Japan

[edit]

Meanwhile, in the mid-80's, Japanese researchers at the central research laboratory of Hitachi Ltd set out to build a NIRS-based brain monitoring system using a pulse of 70-picosecond rays. This effort came into light when the team, along with their leading expert, Dr Hideaki Koizumi (小泉 英明), held an open symposium to announce the principle of "Optical Topography" in January 1995. In fact, the term "Optical Topography" derives from the concept of using light on "2-Dimensional mapping combined with 1-Dimensional information", or topography. The idea had been successfully implemented in launching their first fNIRS (or Optical Topography, as they call it) device based on Frequency Domain in 2001: Hitachi ETG-100. Later, Harumi Oishi (大石 晴美), a PhD-to-be at Nagoya University, published her doctoral dissertation in 2003 with the subject of "language learners' cortical activation patterns measured by ETG-100" under the supervision of Professor Toru Kinoshita (木下 微)—presenting a new prospect on the use of fNIRS. The company has been advancing the ETG series ever since.

## Spectroscopic techniques

[edit]

Currently, there are three modalities of fNIR spectroscopy:

1\. Continuous wave

2\. Frequency domain

3\. Time-domain

### Continuous wave

[edit]

Continuous wave (CW) system uses light sources with constant frequency and amplitude. In fact, to measure absolute changes in HbO concentration with the mBLL, we need to know photon path-length. However, CW-fNIRS does not provide any knowledge of photon path-length, so changes in HbO concentration are relative to an unknown path-length. Many CW-fNIRS commercial systems use estimations of photon path-length derived from computerized Monte-Carlo simulations and physical models, to approximate absolute quantification of hemoglobin concentrations.

OD = ln ⁡ ( I 0 / I ) = ϵ ⋅ [ X ] ⋅ l ⋅ DPF + G {\displaystyle {\text{OD}}=\operatorname {ln} (I_{0}/I)=\epsilon \cdot [X]\cdot l\cdot {\text{DPF}}+G}

Where  OD {\displaystyle {\text{OD}}} is the optical density or attenuation,  I 0 {\displaystyle I_{0}} is emitted light intensity,  I {\displaystyle I} is measured light intensity,  ϵ {\displaystyle \epsilon } is the attenuation coefficient,  [ X ] {\displaystyle [X]} is the chromophore concentration,  l {\displaystyle l} is the distance between source and detector and  DPF {\displaystyle {\text{DPF}}} is the differential path length factor, and  G {\displaystyle G} is a geometric factor associated with scattering.

When the attenuation coefficients  ϵ {\displaystyle \epsilon } are known, constant scattering loss is assumed, and the measurements are treated differentially in time, the equation reduces to:

Δ [ X ] = Δ OD ϵ d {\displaystyle \Delta [X]=\Delta {\frac {\text{OD}}{\epsilon d}}}

Where  d {\displaystyle d} is the total corrected photon path-length.

Using a dual wavelength system, measurements for HbO2 and Hb can be solved from the matrix equation:[9]

( Δ OD λ 1 Δ OD λ 2 ) = ( ϵ λ 1 Hb d ϵ λ 1 HbO 2 d ϵ λ 2 Hb d ϵ λ 2 HbO 2 d ) ( Δ [ X ] Hb Δ [ X ] HbO 2 ) {\displaystyle {\begin{pmatrix}\Delta {\text{OD}}_{\lambda _{1}}\\\\\Delta {\text{OD}}_{\lambda _{2}}\end{pmatrix}}={\begin{pmatrix}\epsilon _{\lambda _{1}}^{\text{Hb}}d&\epsilon _{\lambda _{1}}^{{\text{HbO}}_{2}}d\\\\\epsilon _{\lambda _{2}}^{\text{Hb}}d&\epsilon _{\lambda _{2}}^{{\text{HbO}}_{2}}d\end{pmatrix}}{\begin{pmatrix}\Delta [X]^{\text{Hb}}\\\\\Delta [X]^{{\text{HbO}}_{2}}\end{pmatrix}}}

Due to their simplicity and cost-effectiveness, CW-fNIRS is by far the most common form of functional NIRS since it is the cheapest to make, applicable with more channels, and ensures a high temporal resolution. However, it does not distinguish between absorption and scattering changes, and cannot measure absolute absorption values: which means that it is only sensitive to relative change in HbO concentration.

Still, the simplicity and cost-effectiveness of CW-based devices prove themselves to be the most favorable for a number of clinical applications: neonatal care, patient monitoring systems, diffuse optical tomography, and so forth. Moreover, thanks to its portability, wireless CW systems have been developed—allowing individuals to be monitored in ambulatory, clinical and sports environments.[10] [11][12]

### Frequency domain

[edit]

Frequency domain (FD) system comprises NIR laser sources which provide an amplitude-modulated sinusoid at frequencies near 100 MHz. FD-fNIRS measures attenuation, phase shift and the average path length of light through the tissue.

Changes in the back-scattered signal's amplitude and phase provide a direct measurement of absorption and scattering coefficients of the tissue, thus obviating the need for information about photon path-length; and from the coefficients we determine the changes in the concentration of hemodynamic parameters.

Because of the need for modulated lasers as well as phasic measurements, FD system-based devices are more technically complex (therefore more expensive and much less portable) than CW-based ones. However, the system is capable of providing absolute concentrations of HbO and HbR.

### Time domain

[edit]

Time domain (TD) system introduces a short NIR pulse with a pulse length usually in the order of picoseconds—around 70 ps. Through time-of-flight measurements, photon path-length may be directly observed by dividing resolved time by the speed of light. Information about hemodynamic changes can be found in the attenuation, decay, and time profile of the back-scattered signal. For this photon-counting technology is introduced, which counts 1 photon for every 100 pulses to maintain linearity. TD-fNIRS does have a slow sampling rate as well as a limited number of wavelengths. Because of the need for a photon-counting device, high-speed detection, and high-speed emitters, time-resolved methods are the most expensive and technically complicated.

TD-based devices have the highest depth sensitivity and are capable of presenting most accurate values of baseline hemoglobin concentration and oxygenation.

## Diffuse correlation spectroscopy

[edit]

Diffuse correlation spectroscopy (DCS) is a non-invasive optical imaging technique that utilizes coherent near-infrared light to measure local microvascular cerebral blood flow by quantifying the temporal light intensity fluctuations generated by dynamic scattering of moving red blood cells. This dynamic scattering from moving cells causes the detected intensity to temporally fluctuate. These fluctuations can be quantified by the temporal intensity autocorrelation curve of a single speckle. The decay of the autocorrelation curve is fitted with the solution of the correlation diffusion equation to obtain an index of cerebral blood flow.[13][14][15][16]

## System design

[edit]

At least two open-source fNIRS models are available online:

* http://www.opennirs.org
* https://openfnirs.org/

## Data analysis software

[edit]

### HOMER3

[edit]

HOMER3 allows users to obtain estimates and maps of brain activation. It is a set of matlab scripts used for analyzing fNIRS data. This set of scripts has evolved since the early 1990s first as the Photon Migration Imaging toolbox, then HOMER1 and HOMER2, and now HOMER3.[17]

### NIRS toolbox

[edit]

This toolbox is a set of Matlab-based tools for the analysis of functional near-infrared spectroscopy (fNIRS). This toolbox defines the +nirs namespace and includes a series of tools for signal processing, display, and statistics of fNIRS data. This toolbox is built around an object-oriented framework of Matlab classes and namespaces.[18]

### AtlasViewer

[edit]

AtlasViewer allows fNIRS data to be visualized on a model of the brain. In addition, it also allows the user to design probes which can eventually be placed onto a subject.[19]

## Application

[edit]

### Brain–computer interface

[edit]

fNIRS has been successfully implemented as a control signal for brain–computer interface systems.[20][21][22][23][24]

### Hypoxia & altitude studies

[edit]

With our constant need for oxygen, our body has developed multiple mechanisms that detect oxygen levels, which in turn can activate appropriate responses to counter hypoxia and generate a higher oxygen supply. Moreover, understanding the physiological mechanism underlying the bodily response to oxygen deprivation is of major importance and NIRS devices have shown to be a great tool in this field of research.[25]

Measurement of brain oxyhemoglobin and deoxyhemoglobin concentration changes at high alltitude induced hypoxia with a portable fNIRS device (PortaLite, Artinis Medical Systems)

### Brain mapping

[edit]

### Functional connectivity

[edit]

fNIRS measurements can be used to calculate a limited degree of functional connectivity. Multi-channel fNIRS measurements create a topographical map of neural activation, whereby temporal correlation between spatially separated events can be analyzed. Functional connectivity is typically assessed in terms correlations between the hemodynamic responses of spatially distinct regions of interest (ROIs). In brain studies, functional connectivity measurements are commonly taken for resting state patient data, as well as data recorded over stimulus paradigms. A study led by Alessandro Crimi team highlighted that the functional connectivity measures obtained with fNIRS measurements are quite different from those obtained via EEG caps. [26]

### Cerebral oximetry

[edit]

NIRS monitoring is helpful in a number of ways. Preterm infants can be monitored reducing cerebral hypoxia and hyperoxia with different patterns of activities.[27] It is an effective aid in Cardiopulmonary bypass, is strongly considered to improve patient outcomes and reduce costs and extended stays.

There are inconclusive results for use of NIRS with patients with traumatic brain injury, so it has been concluded that it should remain a research tool.

### Diffuse optical tomography

[edit]

Diffuse optical tomography is the 3D version of Diffuse optical imaging. Diffuse optical images are obtained using NIRS or fluorescence-based methods. These images can be used to develop a 3D volumetric model which is known as the Diffuse Optical Tomography.[28]

10-20 system

### fNIRS cap

[edit]

fNIRS electrode locations can be defined using a variety of layouts, including names and locations that are specified by the International 10–20 system as well as other layouts that are specifically optimized to maintain a consistent 30mm distance between each location. In addition to the standard positions of electrodes, short separation channels can be added. Short separation channels allow the measurement of scalp signals. Since the short separation channels measure the signal coming from the scalp, they allow the removal of the signal of superficial layers. This leaves behind the actual brain response. Short separation channel detectors are usually placed 8mm away from a source. They do not need to be in a specific direction or in the same direction as a detector.[29]

### Functional neuroimaging

[edit]

The use of fNIRS as a functional neuroimaging method relies on the principle of neuro-vascular coupling also known as the haemodynamic response or blood-oxygen-level dependent (BOLD) response. This principle also forms the core of fMRI techniques. Through neuro-vascular coupling, neuronal activity is linked to related changes in localized cerebral blood flow. fNIRS and fMRI are sensitive to similar physiologic changes and are often comparative methods. Studies relating fMRI and fNIRS show highly correlated results in cognitive tasks. fNIRS has several advantages in cost and portability over fMRI, but cannot be used to measure cortical activity more than 4 cm deep due to limitations in light emitter power and has more limited spatial resolution. fNIRS includes the use of diffuse optical tomography (DOT/NIRDOT) for functional purposes. Multiplexing fNIRS channels can allow 2D topographic functional maps of brain activity (e.g. with Hitachi ETG-4000, Artinis Oxymon, NIRx NIRScout, etc.) while using multiple emitter spacings may be used to build 3D tomographic maps.

fNIRS hyperscanning with two violinists

### Hyperscanning

[edit]

See also: Neural synchrony § Hyperscanning

Hyperscanning involves two or more brains monitored simultaneously to investigate interpersonal (across-brains) neural correlates in various social situations, which proves fNIRS to be a suitable modality for investigating live brain-to-brain social interactions.[30]

### Virtual and augmented reality

[edit]

Modern fNIRS systems are combined with virtual or augmented reality in studies on brain-computer interfaces,[31] neurorehabilitation[32] or social perception.[33]

Mobile and wireless fNIRS and EEG systems synchronized with all-in-one head mounted display (PhotonCap, Cortivision)

### Music and the brain

[edit]

fNIRS with a pianist

fNIRS can be used to monitor musicians' brain activity while playing musical instruments.[34][35][36][37]

## Advantages and disadvantages

[edit]

The advantages of fNIRS are, among other things: noninvasiveness, low-cost modalities, perfect safety, high temporal resolution, compatibility with other imaging modalities, and multiple hemodynamic biomarkers.

Disadvantages of fNIRS include: low brain sensitivity given that it can only detect changes on the cortical surface and low spatial resolution. Importantly, the signal is sensitive to hair and skin pigment differences, making it difficult to do between-subject designs. Dense or extremely curly hair may prohibit placement of electrodes close to the scalp, limiting the ability to use the technique with all individuals.

## Future directions

[edit]

fNIRS devices have many attractive features - they are small, lightweight, portable and wearable. They have the potential to be used in clinics, global health,[38] a natural environment, and as a health tracker. Nevertheless, the negatives are salient and must be considered when interpreting the signal.

Now there are fully wireless research grade fNIRS systems in the market.[39]

## fNIRS compared with other neuroimaging techniques

[edit]

When comparing and contrasting neuroimaging devices it is important to look at the temporal resolution, spatial resolution, and the degree of immobility. In particular, EEG (electroencephalograph) and MEG (magnetoencephalography) have high temporal resolution, but a low spatial resolution. EEG also has a higher degree of mobility than MEG has. When looking at fNIRS, they are similar to an EEG. They have a high degree of mobility as well as temporal resolution, and they have low spatial resolution. PET scans and fMRIs are grouped together, however they are distinctly different from the other neuroimaging scans. They have a high degree of immobility, medium/high spatial resolution, and a low temporal resolution. All of these neuroimaging scans have important characteristics and are valuable, however they have distinct characteristics.

Among all other facts, what makes fNIRS a special point of interest is that it is compatible with some of these modalities, including: MRI, EEG, and MEG.

## See also

[edit]

* Near-infrared spectroscopy
* Diffuse optical tomography[40]
* Functional neuroimaging
* Cognitive neuroscience[29]
* The Society for Functional Near Infrared Society (external link)
* Global fNIRS (external Link)
* An example of a mobile fNIRS system designed for studies in VR environments
* Soterix Medical fNIRS
* Cortech Solutions fNIRS
* Neural synchrony

## References

[edit]

1. ^ Ferrari, Marco; Quaresima, Valentina (November 2012). "A brief review on the history of human functional near-infrared spectroscopy (fNIRS) development and fields of application". NeuroImage. 63 (2): 921–935\. doi:10.1016/j.neuroimage.2012.03.049. PMID 22510258. S2CID 18367840.921-935&rft.date=2012-11&rft_id=https://api.semanticscholar.org/CorpusID:18367840#id-name=S2CID&rft_id=info:pmid/22510258&rft_id=info:doi/10.1016/j.neuroimage.2012.03.049&rft.aulast=Ferrari&rft.aufirst=Marco&rft.au=Quaresima,+Valentina&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
2. ^ Cui, Xu; Bray, Signe; Bryant, Daniel M.; Glover, Gary H.; Reiss, Allan L. (February 2011). "A quantitative comparison of NIRS and fMRI across multiple cognitive tasks". NeuroImage. 54 (4): 2808–2821\. doi:10.1016/j.neuroimage.2010.10.069. PMC 3021967. PMID 21047559.2808-2821&rft.date=2011-02&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3021967#id-name=PMC&rft_id=info:pmid/21047559&rft_id=info:doi/10.1016/j.neuroimage.2010.10.069&rft.aulast=Cui&rft.aufirst=Xu&rft.au=Bray,+Signe&rft.au=Bryant,+Daniel+M.&rft.au=Glover,+Gary+H.&rft.au=Reiss,+Allan+L.&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3021967&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
3. ^ Villringer, A.; Chance, B. (1997). "Non-invasive optical spectroscopy and imaging of human brain function". Trends in Neurosciences. 20 (10): 435–442\. doi:10.1016/S0166-2236(97)01132-6. PMID 9347608. S2CID 18077839.435-442&rft.date=1997&rft_id=https://api.semanticscholar.org/CorpusID:18077839#id-name=S2CID&rft_id=info:pmid/9347608&rft_id=info:doi/10.1016/S0166-2236(97)01132-6&rft.aulast=Villringer&rft.aufirst=A.&rft.au=Chance,+B.&rft_id=https://doi.org/10.1016%2FS0166-2236%2897%2901132-6&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
4. ^ Li, Ting; Gong, Hui; Luo, Qingming (1 April 2011). "Visualization of light propagation in visible Chinese human head for functional near-infrared spectroscopy". Journal of Biomedical Optics. 16 (4): 045001. Bibcode:2011JBO....16d5001L. doi:10.1117/1.3567085. PMID 21529068.
5. ^ Kohno, Satoru; Miyai, Ichiro; Seiyama, Akitoshi; Oda, Ichiro; Ishikawa, Akihiro; Tsuneishi, Shoichi; Amita, Takashi; Shimizu, Koji (2007). "Removal of the skin blood flow artifact in functional near-infrared spectroscopic imaging data through independent component analysis". Journal of Biomedical Optics. 12 (6): 062111. Bibcode:2007JBO....12f2111K. doi:10.1117/1.2814249. PMID 18163814.
6. ^ Brigadoi, Sabrina; Cooper, Robert J. (26 May 2015). "How short is short? Optimum source–detector distance for short-separation channels in functional near-infrared spectroscopy". Neurophotonics. 2 (2): 025005. doi:10.1117/1.NPh.2.2.025005. PMC 4478880. PMID 26158009.
7. ^ Modified Beer Lambert Law, archived from the original on 2021-12-21, retrieved 2020-03-26
8. ^ Jöbsis (1997). "Noninvasive, infrared monitoring of cerebral and myocardial oxygen sufficiency and circulatory parameters". Science. 198 (4323): 1264–1267\. doi:10.1126/science.929199. PMID 929199.1264-1267&rft.date=1997&rft_id=info:doi/10.1126/science.929199&rft_id=info:pmid/929199&rft.au=Jöbsis&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
9. ^ Ayaz, Hasan; Shewokis, Patricia A.; Curtin, Adrian; Izzetoglu, Meltem; Izzetoglu, Kurtulus; Onaral, Banu (8 October 2011). "Using MazeSuite and Functional Near Infrared Spectroscopy to Study Learning in Spatial Navigation". Journal of Visualized Experiments (56): 3443. doi:10.3791/3443. PMC 3227178. PMID 22005455.
10. ^ Piper, Sophie K.; Krueger, Arne; Koch, Stefan P.; Mehnert, Jan; Habermehl, Christina; Steinbrink, Jens; Obrig, Hellmuth; Schmitz, Christoph H. (January 2014). "A wearable multi-channel fNIRS system for brain imaging in freely moving subjects". NeuroImage. 85 (1): 64–71\. doi:10.1016/j.neuroimage.2013.06.062. PMC 3859838. PMID 23810973.64-71&rft.date=2014-01&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3859838#id-name=PMC&rft_id=info:pmid/23810973&rft_id=info:doi/10.1016/j.neuroimage.2013.06.062&rft.aulast=Piper&rft.aufirst=Sophie+K.&rft.au=Krueger,+Arne&rft.au=Koch,+Stefan+P.&rft.au=Mehnert,+Jan&rft.au=Habermehl,+Christina&rft.au=Steinbrink,+Jens&rft.au=Obrig,+Hellmuth&rft.au=Schmitz,+Christoph+H.&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3859838&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
11. ^ Curtin, Adrian; Ayaz, Hasan (October 2018). "The Age of Neuroergonomics: Towards Ubiquitous and Continuous Measurement of Brain Function with fNIRS: The age of neuroergonomics and fNIRS". Japanese Psychological Research. 60 (4): 374–386\. doi:10.1111/jpr.12227.374-386&rft.date=2018-10&rft_id=info:doi/10.1111/jpr.12227&rft.aulast=Curtin&rft.aufirst=Adrian&rft.au=Ayaz,+Hasan&rft_id=https://doi.org/10.1111%2Fjpr.12227&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
12. ^ Quaresima, Valentina; Ferrari, Marco (January 2019). "Functional Near-Infrared Spectroscopy (fNIRS) for Assessing Cerebral Cortex Function During Human Behavior in Natural/Social Situations: A Concise Review". Organizational Research Methods. 22 (1): 46–68\. doi:10.1177/1094428116658959. S2CID 148042299.46-68&rft.date=2019-01&rft_id=info:doi/10.1177/1094428116658959&rft_id=https://api.semanticscholar.org/CorpusID:148042299#id-name=S2CID&rft.aulast=Quaresima&rft.aufirst=Valentina&rft.au=Ferrari,+Marco&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
13. ^ Durduran, T.; Yodh, A. G. (2013). "Diffuse correlation spectroscopy for non-invasive, micro-vascular cerebral blood flow measurement". NeuroImage. 85 (1): 51–63\. doi:10.1016/j.neuroimage.2013.06.017. PMC 3991554. PMID 23770408.51-63&rft.date=2013&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3991554#id-name=PMC&rft_id=info:pmid/23770408&rft_id=info:doi/10.1016/j.neuroimage.2013.06.017&rft.aulast=Durduran&rft.aufirst=T.&rft.au=Yodh,+A.+G.&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3991554&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
14. ^ Sutin, Jason; Zimmerman, Bernhard; Tyulmankov, Danil; Tamborini, Davide; Wu, Kuan Cheng; Selb, Juliette; Gulinatti, Angelo; Rech, Ivan; Tosi, Alberto; Boas, David A.; Franceschini, Maria Angela (20 September 2016). "Time-domain diffuse correlation spectroscopy". Optica. 3 (9): 1006–1013\. Bibcode:2016Optic...3.1006S. doi:10.1364/OPTICA.3.001006. PMC 5166986. PMID 28008417.1006-1013&rft.date=2016-09-20&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC5166986#id-name=PMC&rft_id=info:pmid/28008417&rft_id=info:doi/10.1364/OPTICA.3.001006&rft_id=info:bibcode/2016Optic...3.1006S&rft.aulast=Sutin&rft.aufirst=Jason&rft.au=Zimmerman,+Bernhard&rft.au=Tyulmankov,+Danil&rft.au=Tamborini,+Davide&rft.au=Wu,+Kuan+Cheng&rft.au=Selb,+Juliette&rft.au=Gulinatti,+Angelo&rft.au=Rech,+Ivan&rft.au=Tosi,+Alberto&rft.au=Boas,+David+A.&rft.au=Franceschini,+Maria+Angela&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC5166986&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
15. ^ Carp, S. A.; Tamborini, D.; Mazumder, D.; Wu, K. C.; Robinson, M. R.; Stephens, K. A.; Shatrovoy, O.; Lue, N.; Ozana, N.; Blackwell, M. H.; Franceschini, M. A. (2020). "Diffuse correlation spectroscopy measurements of blood flow using 1064 nm light". Journal of Biomedical Optics. 25 (9): 097003. Bibcode:2020JBO....25i7003C. doi:10.1117/1.JBO.25.9.097003. PMC 7522668. PMID 32996299.
16. ^ Buckley, Erin M.; Parthasarathy, Ashwin B.; Grant, P. Ellen; Yodh, Arjun G.; Franceschini, Maria Angela (2014). "Diffuse correlation spectroscopy for measurement of cerebral blood flow: Future prospects". Neurophotonics. 1 (1): 011009. doi:10.1117/1.NPh.1.1.011009. PMC 4292799. PMID 25593978. S2CID 13208535.
17. ^ "HOMER2". HOMER2. Retrieved 2019-11-26.
18. ^ Santosa, H., Zhai, X., Fishburn, F., & Huppert, T. (2018). The NIRS Brain AnalyzIR Toolbox. Algorithms, 11(5), 73.
19. ^ Aasted, Christopher M.; Yücel, Meryem A.; Cooper, Robert J.; Dubb, Jay; Tsuzuki, Daisuke; Becerra, Lino; Petkov, Mike P.; Borsook, David; Dan, Ippeita; Boas, David A. (5 May 2015). "Anatomical guidance for functional near-infrared spectroscopy: AtlasViewer tutorial". Neurophotonics. 2 (2): 020801. doi:10.1117/1.NPh.2.2.020801. PMC 4478785. PMID 26157991.
20. ^ Ayaz, H.; Shewokis, P. A.; Bunce, S.; Onaral, B. (2011). "An optical brain computer interface for environmental control". Annual International Conference of the IEEE Engineering in Medicine and Biology Society. IEEE Engineering in Medicine and Biology Society. Annual International Conference. Vol. 2011\. pp. 6327–6330\. doi:10.1109/IEMBS.2011.6091561. ISBN 978-1-4577-1589-1. PMID 22255785. S2CID 4951918.6327-6330&rft.date=2011&rft_id=info:doi/10.1109/IEMBS.2011.6091561&rft_id=https://api.semanticscholar.org/CorpusID:4951918#id-name=S2CID&rft_id=info:pmid/22255785&rft.isbn=978-1-4577-1589-1&rft.aulast=Ayaz&rft.aufirst=H.&rft.au=Shewokis,+P.+A.&rft.au=Bunce,+S.&rft.au=Onaral,+B.&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
21. ^ Coyle, Shirley M; Ward, Tomás E; Markham, Charles M (September 2007). "Brain–computer interface using a simplified functional near-infrared spectroscopy system" (PDF). Journal of Neural Engineering. 4 (3): 219–226\. Bibcode:2007JNEng...4..219C. doi:10.1088/1741-2560/4/3/007. PMID 17873424. S2CID 18723855.219-226&rft.date=2007-09&rft_id=info:doi/10.1088/1741-2560/4/3/007&rft_id=https://api.semanticscholar.org/CorpusID:18723855#id-name=S2CID&rft_id=info:pmid/17873424&rft_id=info:bibcode/2007JNEng...4..219C&rft.aulast=Coyle&rft.aufirst=Shirley+M&rft.au=Ward,+Tomás+E&rft.au=Markham,+Charles+M&rft_id=https://mural.maynoothuniversity.ie/15548/1/CM_brain-computer.pdf&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
22. ^ Sitaram, Ranganatha; Zhang, Haihong; Guan, Cuntai; Thulasidas, Manoj; Hoshi, Yoko; Ishikawa, Akihiro; Shimizu, Koji; Birbaumer, Niels (February 2007). "Temporal classification of multichannel near-infrared spectroscopy signals of motor imagery for developing a brain–computer interface". NeuroImage. 34 (4): 1416–1427\. doi:10.1016/j.neuroimage.2006.11.005. PMID 17196832. S2CID 15471179.1416-1427&rft.date=2007-02&rft_id=https://api.semanticscholar.org/CorpusID:15471179#id-name=S2CID&rft_id=info:pmid/17196832&rft_id=info:doi/10.1016/j.neuroimage.2006.11.005&rft.aulast=Sitaram&rft.aufirst=Ranganatha&rft.au=Zhang,+Haihong&rft.au=Guan,+Cuntai&rft.au=Thulasidas,+Manoj&rft.au=Hoshi,+Yoko&rft.au=Ishikawa,+Akihiro&rft.au=Shimizu,+Koji&rft.au=Birbaumer,+Niels&rft_id=https://ink.library.smu.edu.sg/sis_research/3493&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
23. ^ Naseer, Noman; Hong, Melissa Jiyoun; Hong, Keum-Shik (February 2014). "Online binary decision decoding using functional near-infrared spectroscopy for the development of brain–computer interface". Experimental Brain Research. 232 (2): 555–564\. doi:10.1007/s00221-013-3764-1. PMID 24258529. S2CID 15250694.555-564&rft.date=2014-02&rft_id=https://api.semanticscholar.org/CorpusID:15250694#id-name=S2CID&rft_id=info:pmid/24258529&rft_id=info:doi/10.1007/s00221-013-3764-1&rft.aulast=Naseer&rft.aufirst=Noman&rft.au=Hong,+Melissa+Jiyoun&rft.au=Hong,+Keum-Shik&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
24. ^ Naseer, Noman; Hong, Keum-Shik (October 2013). "Classification of functional near-infrared spectroscopy signals corresponding to the right- and left-wrist motor imagery for development of a brain–computer interface". Neuroscience Letters. 553: 84–89\. doi:10.1016/j.neulet.2013.08.021. PMID 23973334. S2CID 220773.84-89&rft.date=2013-10&rft_id=https://api.semanticscholar.org/CorpusID:220773#id-name=S2CID&rft_id=info:pmid/23973334&rft_id=info:doi/10.1016/j.neulet.2013.08.021&rft.aulast=Naseer&rft.aufirst=Noman&rft.au=Hong,+Keum-Shik&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
25. ^ Shaw, Keely; Singh, Jyotpal; Sirant, Luke; Neary, J. Patrick; Chilibeck, Philip D. (November 2020). "Effect of Dark Chocolate Supplementation on Tissue Oxygenation, Metabolism, and Performance in Trained Cyclists at Altitude". International Journal of Sport Nutrition and Exercise Metabolism. 30 (6): 420–426\. doi:10.1123/ijsnem.2020-0051. PMID 32916656. S2CID 221635672.420-426&rft.date=2020-11&rft_id=https://api.semanticscholar.org/CorpusID:221635672#id-name=S2CID&rft_id=info:pmid/32916656&rft_id=info:doi/10.1123/ijsnem.2020-0051&rft.aulast=Shaw&rft.aufirst=Keely&rft.au=Singh,+Jyotpal&rft.au=Sirant,+Luke&rft.au=Neary,+J.+Patrick&rft.au=Chilibeck,+Philip+D.&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
26. ^ Blanco, R; Koba, C; Crimi, A (2024). "Investigating the interaction between EEG and fNIRS: a multimodal network analysis of brain connectivity". Journal of Computational Science. 82: 102416. doi:10.1016/j.jocs.2024.102416.
27. ^ Rahimpour, Ali; Noubari, Hosein Ahmadi; Kazemian, Mohammad (2018). "A case-study of NIRS application for infant cerebral hemodynamic monitoring: A report of data analysis for feature extraction and infant classification into healthy and unhealthy". Informatics in Medicine Unlocked. 11: 44–50\. doi:10.1016/j.imu.2018.04.001.44-50&rft.date=2018&rft_id=info:doi/10.1016/j.imu.2018.04.001&rft.aulast=Rahimpour&rft.aufirst=Ali&rft.au=Noubari,+Hosein+Ahmadi&rft.au=Kazemian,+Mohammad&rft_id=https://doi.org/10.1016%2Fj.imu.2018.04.001&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
28. ^ Durduran, T.; Choe, R.; Baker, W. B.; Yodh, A. G. (July 2010). "Diffuse Optics for Tissue Monitoring and Tomography". Reports on Progress in Physics. 73 (7): 076701. Bibcode:2010RPPh...73g6701D. doi:10.1088/0034-4885/73/7/076701. PMC 4482362. PMID 26120204.
29. ^ a b Yücel, Meryem A.; Selb, Juliette; Aasted, Christopher M.; Petkov, Mike P.; Becerra, Lino; Borsook, David; Boas, David A. (11 September 2015). "Short separation regression improves statistical significance and better localizes the hemodynamic response obtained by near-infrared spectroscopy for tasks with differing autonomic responses". Neurophotonics. 2 (3): 035005. doi:10.1117/1.NPh.2.3.035005. PMC 4717232. PMID 26835480.
30. ^ mari (2018-02-04). "fNIRS Hyperscanning: A door to real-world social neuroscience research". The Society for functional Near Infrared Spectroscopy. Retrieved 2020-03-26.
31. ^ Piper, Sophie K.; Krueger, Arne; Koch, Stefan P.; Mehnert, Jan; Habermehl, Christina; Steinbrink, Jens; Obrig, Hellmuth; Schmitz, Christoph H. (15 January 2014). "A wearable multi-channel fNIRS system for brain imaging in freely moving subjects". NeuroImage. 85 (1): 64–71\. doi:10.1016/j.neuroimage.2013.06.062. PMC 3859838. PMID 23810973.64-71&rft.date=2014-01-15&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3859838#id-name=PMC&rft_id=info:pmid/23810973&rft_id=info:doi/10.1016/j.neuroimage.2013.06.062&rft.aulast=Piper&rft.aufirst=Sophie+K.&rft.au=Krueger,+Arne&rft.au=Koch,+Stefan+P.&rft.au=Mehnert,+Jan&rft.au=Habermehl,+Christina&rft.au=Steinbrink,+Jens&rft.au=Obrig,+Hellmuth&rft.au=Schmitz,+Christoph+H.&rft_id=https://www.ncbi.nlm.nih.gov/pmc/articles/PMC3859838&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
32. ^ Holper, Lisa; Muehlemann, Thomas; Scholkmann, Felix; Eng, Kynan; Kiper, Daniel; Wolf, Martin (December 2010). "Testing the potential of a virtual reality neurorehabilitation system during performance of observation, imagery and imitation of motor actions recorded by wireless functional near-infrared spectroscopy (fNIRS)". Journal of NeuroEngineering and Rehabilitation. 7 (1): 57. doi:10.1186/1743-0003-7-57. PMC 3014953. PMID 21122154.
33. ^ Kim, Gyoung; Buntain, Noah; Hirshfield, Leanne; Costa, Mark R.; Chock, T. Makana (2019). "Processing Racial Stereotypes in Virtual Reality: An Exploratory Study Using Functional Near-Infrared Spectroscopy (FNIRS)". Augmented Cognition. Lecture Notes in Computer Science. Vol. 11580\. pp. 407–417\. doi:10.1007/978-3-030-22419-6_29. ISBN 978-3-030-22418-9. S2CID 195891659.407-417&rft.date=2019&rft_id=https://api.semanticscholar.org/CorpusID:195891659#id-name=S2CID&rft_id=info:doi/10.1007/978-3-030-22419-6_29&rft.isbn=978-3-030-22418-9&rft.aulast=Kim&rft.aufirst=Gyoung&rft.au=Buntain,+Noah&rft.au=Hirshfield,+Leanne&rft.au=Costa,+Mark+R.&rft.au=Chock,+T.+Makana&rfr_id=info:sid/en.wikipedia.org:Functional+near-infrared+spectroscopy" class="Z3988">
34. ^ "YouTube". www.youtube.com. Archived from the original on 2021-12-21. Retrieved 2020-03-26.
35. ^ fNIRS of playing piano, archived from the original on 2021-12-21, retrieved 2020-03-26
36. ^ fNIRS of Observation, archived from the original on 2021-12-21, retrieved 2020-03-26
37. ^ fNIRS of Imagery, archived from the original on 2021-12-21, retrieved 2020-03-26
38. ^ Lloyd-Fox, Sarah; Papademetriou, M.; Darboe, M. K.; Everdell, N. L.; Wegmuller, R.; Prentice, A. M.; Moore, S. E.; Elwell, C. E. (22 April 2014). "Functional near infrared spectroscopy (fNIRS) to assess cognitive function in infants in rural Africa". Scientific Reports. 4 (1): 4740. Bibcode:2014NatSR...4E4740L. doi:10.1038/srep04740. PMC 5381189. PMID 24751935. S2CID 8522984.
39. ^ Shin, Jaeyoung; Kwon, Jinuk; Choi, Jongkwan; Im, Chang-Hwan (29 November 2017). "Performance enhancement of a brain-computer interface using high-density multi-distance NIRS". Scientific Reports. 7 (1): 16545. Bibcode:2017NatSR...716545S. doi:10.1038/s41598-017-16639-0. PMC 5707382. PMID 29185494.
40. ^ "NIRx | fNIRS Systems | NIRS Devices". NIRx Medical Technologies. Retrieved 2019-11-26.