LOPES (exoskeleton)

This article has multiple issues. Please help improve it or discuss these issues on the talk page. (Learn how and when to remove these messages)

This article contains promotional content. Please help improve it by removing promotional language and inappropriate external links, and by adding encyclopedic text written from a neutral point of view. (June 2024) (Learn how and when to remove this message)

|

This article has no lead section. Please improve this article by adding one in your own words. (June 2024) (Learn how and when to remove this message)

|

This article needs additional citations for verification. Please help improve this article by adding citations to reliable sources. Unsourced material may be challenged and removed.
Find sources: "LOPES" exoskeleton – news · newspapers · books · scholar · JSTOR (June 2024) (Learn how and when to remove this message)

|

(Learn how and when to remove this message)

The goal of the LOPES project (LOwer-extremity Powered ExoSkeleton) is to design and implement a gait rehabilitation robot for treadmill training. The target group consists of people who have had a stroke and have impaired motor control. The main goals of LOPES are:

* Reduction of the physical load on the therapist / patient;
* More efficient gait training for stroke patients;
* Selective support of gait functions;
* Therapist stays in charge of high-level decisions.

The mechanical construction should offer assistance in leg movements in the forward direction and in keeping lateral balance. Within the LOPES project, it has been decided to realize this by connecting the limbs of the patient to an exoskeleton so that robot and patient move in parallel. Most gait rehabilitation robots that are currently being developed[1][2] focus on the support of the entire gait cycle as a single unit. These robots use joint trajectories of the entire gait cycle and offer a uniform (more or less) stiff control along this trajectory. This means that the patient receives support in gait phases where support is necessary but also in phases where support isn't necessary. Studies have been done on an exoskeleton [3] that propose adaptive control methods which minimize the interaction forces with the patient with respect to an adaptable reference pattern, but these still control the entire gait cycle. Studies have also shown that walking with the Lokomat requires significantly less energy than normal walking [4][full citation needed]. LOPES aims to support and not take over those tasks that the patient is unable to perform without help using an impedance control scheme. This will lead to a more active participation from the patient's side. The tradeoff for more active walking will likely be a smaller overall distance during therapy sessions. The implication of selective function support is that the robot will have two extreme modes in which it should be able to function, these are:

* Patient in charge: The goal of the robot is to minimize the interaction forces between the patient and the robot in order for the patient to walk freely without feeling the robot. This mode will be active mostly for the non-paretic side of the patient and during those phases of the walking cycle that the robot does not need to assist.
* Robot in charge: The goal of this mode is to take control of the patient. The robot will take over the functions which the patient is unable to perform. The robot will most likely operate somewhere between these extremes offering some support at those phases when it is needed to guide the patient towards desirable behavior.

The first prototype has been completed. This prototype has 8 actuated DOF (series elastic actuation[5]) following the design.[6] Clinical evaluations will be done in the course of 2007.[needs update][citation needed]

* Medicine portal

## References

[edit]

1. ^ Colombo, Gery; Jörg, Matthias; Schreier, R; Dietz, Volker (2000). "Treadmill training of paraplegic patients using a robotic orthosis". Journal of Rehabilitation Research and Development. 37 (6): 693–700\. PMID 11321005.693-700&rft.date=2000&rft_id=info:pmid/11321005&rft.aulast=Colombo&rft.aufirst=Gery&rft.au=Jörg,+Matthias&rft.au=Schreier,+R&rft.au=Dietz,+Volker&rfr_id=info:sid/en.wikipedia.org:LOPES+(exoskeleton)" class="Z3988">
2. ^ Schmidt, H; Sorowka, D; Hesse, Stefan; Bernhardt, R (2003). "Development of a Robotic Walking Simulator for Gait Rehabilitation". Biomedizinische Technik. 48 (10): 281–286\. doi:10.1515/bmte.2003.48.10.281. PMID 14606269. S2CID 29801731.281-286&rft.date=2003&rft_id=https://api.semanticscholar.org/CorpusID:29801731#id-name=S2CID&rft_id=info:pmid/14606269&rft_id=info:doi/10.1515/bmte.2003.48.10.281&rft.aulast=Schmidt&rft.aufirst=H&rft.au=Sorowka,+D&rft.au=Hesse,+Stefan&rft.au=Bernhardt,+R&rfr_id=info:sid/en.wikipedia.org:LOPES+(exoskeleton)" class="Z3988">
3. ^ Jezernik, Sašo; Colombo, Gery; Morari, Manfred (2004). "Automatic Gait-Pattern Adaptation Algorithms for Rehabilitation With a 4-DOF Robotic Orthosis". IEEE Transactions on Robotics and Automation. 20 (3): 574. doi:10.1109/TRA.2004.825515. S2CID 14766299.
4. ^ C. Krewer, F.M., B. Husemann, S. Heller, J. Quintern, E. Koenig. Energy expenditure of hemiparetic patients and healthy subjects: walking in a lokomat vs. on a treadmill. in Evidence-Based Medicine in Neurorehabilitation. 2004. Zürich.
5. ^ "Series Elastic Actuators Description". Archived from the original on 10 February 2012.
6. ^ Veneman, J.F.; Ekkelenkamp, R.; Kruidhof, R.; Van Der Helm, F.C.T.; Van Der Kooij, H. (2005), "Design of a Series Elastic and Bowdencable-Based Actuation System for Use as Torque-Actuator in Exoskeleton-Type Training Robots", 9th International Conference on Rehabilitation Robotics, 2005. ICORR 2005, pp. 496–499, doi:10.1109/ICORR.2005.1501150, ISBN 0-7803-9003-2, S2CID 18471782496-499&rft.date=2005&rft_id=https://api.semanticscholar.org/CorpusID:18471782#id-name=S2CID&rft_id=info:doi/10.1109/ICORR.2005.1501150&rft.isbn=0-7803-9003-2&rft.aulast=Veneman&rft.aufirst=J.F.&rft.au=Ekkelenkamp,+R.&rft.au=Kruidhof,+R.&rft.au=Van+Der+Helm,+F.C.T.&rft.au=Van+Der+Kooij,+H.&rft_id=https://research.utwente.nl/en/publications/d89adad3-b3f1-42ef-b259-b0e1b698e8d6&rfr_id=info:sid/en.wikipedia.org:LOPES+(exoskeleton)" class="Z3988">

## External links

[edit]

* Official website