Modes of mechanical ventilation

Methods of inspiratory support

This article needs additional citations for verification. Please help improve this article by adding citations to reliable sources. Unsourced material may be challenged and removed.
Find sources: "Modes of mechanical ventilation" – news · newspapers · books · scholar · JSTOR (April 2020) (Learn how and when to remove this message)

|

Modes of mechanical ventilation are one of the most important aspects of the usage of mechanical ventilation. The mode refers to the method of inspiratory support. In general, mode selection is based on clinician familiarity and institutional preferences, since there is a paucity of evidence indicating that the mode affects clinical outcome. The most frequently used forms of volume-limited mechanical ventilation are intermittent mandatory ventilation (IMV) and continuous mandatory ventilation (CMV).[1]

There have been substantial changes in the nomenclature of mechanical ventilation over the years, but more recently it has become standardized by many respirology and pulmonology groups.[2][3]

Writing a mode is most proper in all capital letters with a dash between the control variable and the strategy (i.e. PC-IMV, or VC-MMV etc.).

## Taxonomy for mechanical ventilation

[edit]

The taxonomy is a logical classification system based on 10 maxims of ventilator design:[4]

### 10 maxims

[edit]

1. A breath is one cycle of positive flow (inspiration) and negative flow (expiration) defined in terms of the flow-time curve. Inspiratory time is defined as the period from the start of positive flow to the start of negative flow. Expiratory time is defined as the period from the start of expiratory flow to the start of inspiratory flow. The flow-time curve is the basis for many variables related to ventilator settings.
2. A breath is assisted if the ventilator does work on the patient. An assisted breath is one for which the ventilator does some portion of the work of breathing. For constant flow inflation, work is defined as inspiratory pressure multiplied by tidal volume. Therefore, an assisted breath is identified as a breath for which airway pressure (displayed on the ventilator) rises above baseline during inspiration. An unassisted breath is one for which the ventilator simply provides the inspiratory flow demanded by the patient and pressure stays constant throughout the breath.
3. A ventilator assists breathing using either pressure control or volume control based on the equation of motion for the respiratory system. Providing assistance means doing work on the patient, which is accomplished by controlling either pressure or volume. A simple mathematical model describing this fact is known as the equation of motion for the passive respiratory system:

> Pressure = (Elastance × Volume) + (Resistance × Flow)

In this equation, pressure, volume, and flow are all continuous functions of time. Pressure is actually a pressure difference across the system (e.g., transrespiratory pressure defined as pressure at the airway opening minus pressure on the body surface). Elastance (defined as the change in pressure divided by the associated change in volume; the reciprocal of compliance) and resistance (defined as a change in pressure divided by the associated change in flow) are parameters assumed to remain constant during a breath.

> Volume control (VC) means that both volume and flow are preset prior to inspiration. In other words, the right hand side of the equation of motion remains constant while pressure changes with changes in elastance and resistance.
> Pressure control (PC) means that inspiratory pressure is preset as either a constant value or it is proportional to the patient's inspiratory effort. In other words, the left-hand side of the equation of motion remains constant while volume and flow change with changes in elastance and resistance.
> Time control (TC) means that, in some rare situations, none of the main variables (pressure, volume, or flow) are preset. In this case only the inspiratory and expiratory times are preset.

4. Breaths are classified by the criteria that trigger (start) and cycle (stop) inspiration. The start of inspiration is called the trigger event. The end of inspiration is called the cycle event.
5. Trigger and cycle events can be initiated by the patient or the machine. Inspiration can be patient triggered or patient cycled by a signal representing inspiratory effort. Inspiration may also be machine triggered or machine cycled by preset ventilator thresholds.

> Patient triggering means starting inspiration based on a patient signal independent of a machine trigger signal. Machine triggering means starting inspiratory flow based on a signal from the ventilator, independent of a patient trigger signal. Patient cycling means ending inspiratory time based on signals representing the patient determined components of the equation of motion, (ie, elastance or resistance and including effects due to inspiratory effort). Flow cycling is a form of patient cycling because the rate of flow decay to the cycle threshold is determined by patient mechanics. Machine cycling means ending inspiratory time independent of signals representing the patient determined components of the equation of motion.

6. Breaths are classified as spontaneous or mandatory based on both the trigger and cycle events. A spontaneous breath is a breath for which the patient both triggers and cycles the breath. A spontaneous breath may occur during a mandatory breath (e.g. Airway Pressure Release Ventilation). A spontaneous breath may be assisted or unassisted. A mandatory breath is a breath for which the machine triggers and/or cycles the breath. A mandatory breath can occur during a spontaneous breath (e.g., High Frequency Jet Ventilation). A mandatory breath is, by definition, assisted.
7. There are 3 breath sequences: Continuous mandatory ventilation (CMV), Intermittent Mandatory Ventilation (IMV), and Continuous Spontaneous Ventilation (CSV). A breath sequence is a particular pattern of spontaneous and/or mandatory breaths. The 3 possible breath sequences are: continuous mandatory ventilation, (CMV, spontaneous breaths are not allowed between mandatory breaths), intermittent mandatory ventilation (IMV, spontaneous breaths may occur between mandatory breaths), and continuous spontaneous ventilation (CSV, all breaths are spontaneous).
8. There are 5 basic ventilatory patterns: VC-CMV, VC-IMV, PC-CMV, PC-IMV, and PC-CSV. The combination VC-CSV is not possible because volume control implies machine cycling and machine cycling makes every breath mandatory, not spontaneous. A sixth pattern, TC-IMV is possible but rare.
9. Within each ventilatory pattern there are several variations that can be distinguished by their targeting scheme(s). A targeting scheme is a description of how the ventilator achieves preset targets. A target is a predetermined goal of ventilator output. Examples of within-breath targets include inspiratory flow or pressure and rise time (set-point targeting), tidal volume (dual targeting) and constant of proportionality between inspiratory pressure and patient effort (servo targeting). Examples of between-breath targets and targeting schemes include average tidal volume (for adaptive targeting), percent minute ventilation (for optimal targeting) and combined PCO2, volume, and frequency values describing a "zone of comfort" (for intelligent targeting, e.g., SmartCarePS or IntelliVent-ASV). The targeting scheme (or combination of targeting schemes) is what distinguishes one ventilatory pattern from another. There are 7 basic targeting schemes that comprise the wide variety seen in different modes of ventilation:

> Set-point: A targeting scheme for which the operator sets all the parameters of the pressure waveform (pressure control modes) or volume and flow waveforms (volume control modes).
> Dual: A targeting scheme that allows the ventilator to switch between volume control and pressure control during a single inspiration.
> Bio-variable: A targeting scheme that allows the ventilator to automatically set the inspiratory pressure or tidal volume randomly to mimic the variability observed during normal breathing.
> Servo: A targeting scheme for which inspiratory pressure is proportional to inspiratory effort.
> Adaptive: A targeting scheme that allows the ventilator to automatically set one target (eg, pressure within a breath) to achieve another target (eg, average tidal volume over several breaths).
> Optimal: A targeting scheme that automatically adjusts the targets of the ventilatory pattern to either minimize or maximize some overall performance characteristic (eg, minimize the work rate done by the ventilatory pattern).
> Intelligent: A targeting scheme that uses artificial intelligence programs such as fuzzy logic, rule based expert systems, and artificial neural networks.

10. A mode of ventilation is classified according to its control variable, breath sequence, and targeting scheme(s). The preceding 9 maxims create a theoretical foundation for a taxonomy of mechanical ventilation. The taxonomy is based on these theoretical constructs and has 4 hierarchical levels:

* Control Variable (Pressure or Volume, for the primary breath)
* Breath Sequence (CMV, IMV, or CSV)
* Primary Breath Targeting Scheme (for CMV or CSV)
* Secondary Breath Targeting Scheme (for IMV)

> The "primary breath" is either the only breath there is (mandatory for CMV and spontaneous for CSV) or it is the mandatory breath in IMV. The targeting schemes can be represented by single, lower case letters: set-point = s, dual = d, servo = r, bio-variable = b, adaptive = a, optimal = o, intelligent = i. A tag is an abbreviation for a mode classification, such as PC-IMVs,s. Compound tags are possible, eg, PC-IMVoi,oi.

### How modes are classified

[edit]

This section contains instructions, advice, or how-to content. Please help rewrite the content so that it is more encyclopedic or move it to Wikiversity, Wikibooks, or Wikivoyage. (May 2025)

|

Step 1: Identify the primary breath control variable. If inspiration starts with a preset inspiratory pressure, or if pressure is proportional to inspiratory effort, then the control variable is pressure. If inspiration starts with a preset tidal volume and inspiratory flow, then the control variable is volume. If neither is true, the control variable is time.

Step 2: Identify the breath sequence. Determine whether trigger and cycle events are patient or machine determined. Then, use this information to determine the breath sequence.

Step 3: Identify the targeting schemes for the primary breaths and (if applicable) secondary breaths.

### Example mode classification is given below

[edit]

Mode Name: A/C Volume Control (Covidien PB 840):[citation needed]

1. Inspiratory volume and flow are preset, so the control variable is volume.
2. Every breath is volume cycled, which is a form of machine cycling. Any breath for which inspiration is machine cycled is classified as a mandatory breath. Hence, the breath sequence is continuous mandatory ventilation.
3. The operator sets all the parameters of the volume and flow waveforms so the targeting scheme is set-point. Thus, the mode is classified as volume control continuous mandatory ventilation with set-point targeting (VC-CMVs).

Mode Name: SIMV Volume Control Plus (Covidien PB 840):[citation needed]

1. The operator sets the tidal volume but not the inspiratory flow. Because setting volume alone (like setting flow alone) is a necessary but not sufficient criterion for volume control, the control variable is pressure.
2. Spontaneous breaths are allowed between mandatory breaths so the breath sequence is IMV[clarification needed].
3. The ventilator adjusts inspiratory pressure between breaths to achieve an average preset tidal volume, so the targeting scheme is adaptive. The mode tag is PC-IMVa,s.

## Descriptions of common modes

[edit]

Mechanical ventilation machines are available with both invasive modes (such as intubation) and non-invasive modes (such as BPAP). Invasive has to do with the insertion of medical devices or tubes internal to the patient, while non-invasive is completely external to the patient, as for example in using a tightly fitting mask or other device that covers the patient's nose and mouth.

### Assist mode, control mode, and assist-control mode

[edit]

A basic distinction in mechanical ventilation is whether each breath is initiated by the patient (assist mode) or by the machine (control mode).[citation needed] Dynamic hybrids of the two (assist-control modes) are also possible, and control mode without assist is now mostly obsolete.[citation needed]

### Airway pressure release ventilation

[edit]

Airway pressure release ventilation graph

Airway pressure release ventilation is a time-cycled alternant between two levels of positive airway pressure, with the main time on the high level and a brief expiratory release to facilitate ventilation.[5][6][7][needs update?]

Airway pressure release ventilation is a very versatile mode of ventilation. Like other modes, it needs to be well understood to use it effectively.[8] The exhalation time (Tlow) is shortened to usually less than one second to maintain alveoli inflation. In the basic sense, this is a continuous pressure with a brief release.

Different perceptions of this mode may exist around the globe. While 'APRV' is common to users in North America, a very similar mode, biphasic positive airway pressure (BIPAP), was introduced in Europe.[9][needs update?] The term APRV has also been used in American journals where, from the ventilation characteristics, BIPAP would have been perfectly good terminology.[10][needs update?] But BiPAP(tm) is a trademark for a noninvasive ventilation mode in a specific ventilator (Respironics Inc.).

Other manufacturers have followed with their own brand names (BILEVEL, DUOPAP, BIVENT). Although similar in modality, these terms describe how a mode is intended to inflate the lung, rather than defining the characteristics of synchronization or the way spontaneous breathing efforts are supported.

Intermittent mandatory ventilation has not always had the synchronized feature, so the division of modes were understood to be SIMV (synchronized) vs IMV (not-synchronized). Since the American Association for Respiratory Care established a nomenclature of mechanical ventilation the "synchronized" part of the title has been dropped and now there is only IMV.

### Mandatory minute ventilation

[edit]

Mandatory minute ventilation (MMV) allows spontaneous breathing with automatic adjustments of mandatory ventilation to the meet the patient's preset minimum minute volume requirement. If the patient maintains the minute volume settings for VT x f, no mandatory breaths are delivered.[citation needed]

If the patient's minute volume is insufficient, mandatory delivery of the preset tidal volume will occur until the minute volume is achieved. The method for monitoring whether or not the patient is meeting the required minute ventilation (VE) differs by ventilator brand and model, but, in general, there is a window of monitored time, and a smaller window checked against the larger window (i.e., in the Dräger Evita® line of mechanical ventilators there is a moving 20-second window, and every 7 seconds the current tidal volume and rate are measured) to decide whether a mechanical breath is needed to maintain the minute ventilation.[citation needed]

MMV is an optimal mode for weaning in neonatal and pediatric populations and has been shown to reduce long-term complications related to mechanical ventilation.[11][needs update?]

### Pressure-regulated volume control

[edit]

Pressure-regulated volume control is an Assist Controlled Ventilation (ACV) based mode. Pressure-regulated volume control utilizes pressure-limited, volume-targeted, time-cycled breaths that can be either ventilator- or patient-initiated.[citation needed]

The peak inspiratory pressure delivered by the ventilator is varied on a breath-to-breath basis to achieve a target tidal volume that is set by the clinician.

For example, if a target tidal volume of 500 mL is set but the ventilator delivers 600 mL, the next breath will be delivered with a lower inspiratory pressure to achieve a lower tidal volume. Though PRVC is regarded as a hybrid mode because of its tidal-volume (VC) settings and pressure-limiting (PC) settings fundamentally PRVC is a pressure-control mode with adaptive targeting.

### Continuous positive airway pressure

[edit]

Continuous positive airway pressure (CPAP) is a non-invasive positive pressure mode of respiratory support. CPAP is a continuous pressure applied to keep the alveoli open and not fully deflate. This mechanism for maintaining inflated alveoli helps increase partial pressure of oxygen in arterial blood, an appropriate increase in CPAP increases the PaO2.[citation needed]

#### Automatic positive airway pressure

[edit]

Automatic positive airway pressure (APAP) is a form of CPAP that automatically tunes the amount of pressure delivered to the patient to the minimum required to maintain an unobstructed airway on a breath-by-breath basis by measuring the resistance in the patient's breathing.[citation needed]

### Bilevel positive airway pressure

[edit]

Bilevel positive airway pressure (BPAP) is a mode used during non-invasive ventilation (NIV). First used in 1988 by Professor Benzer in Austria,[12] it delivers a preset inspiratory positive airway pressure (IPAP) and expiratory positive airway pressure (EPAP). BPAP can be described as a Continuous Positive Airway Pressure system with a time-cycle change of the applied CPAP level.[13][needs update?]

CPAP/APAP, BPAP, and other non-invasive ventilation modes have been shown to be effective management tools for chronic obstructive pulmonary disease, acute respiratory failure, sleep apnea, etc.[14][needs update?]

Often BPAP is incorrectly referred to as "BiPAP". BiPAP is the name of a portable ventilator manufactured by Respironics Corporation; it is just one of many ventilators that can deliver BPAP.[citation needed]

#### Medical uses

[edit]

BPAP has been shown to be useful in reducing mortality and reducing the need for endotracheal intubation when used in people with chronic obstructive pulmonary disease (COPD).[15]

### High-frequency ventilation (Active)

[edit]

The term active refers to the ventilator's forced expiratory system. In a HFV-A scenario, the ventilator uses pressure to apply an inspiratory breath and then applies an opposite pressure to force an expiratory breath. In high-frequency oscillatory ventilation (sometimes abbreviated HFOV) the oscillation bellows and piston force positive pressure in and apply negative pressure to force an expiration.[16][needs update?]

### High-frequency ventilation (Passive)

[edit]

The term passive refers to the ventilator's non-forced expiratory system. In a HFV-P scenario, the ventilator uses pressure to apply an inspiratory breath and then returns to atmospheric pressure to allow for a passive expiration.[citation needed]

This is seen in High-Frequency Jet Ventilation, sometimes abbreviated HFJV. Also categorized under High Frequency Ventilation is High Frequency Percussive Ventilation, sometimes abbreviated HFPV. With HFPV it utilizes an open circuit to deliver its subtidal volumes by way of the patient interface known as the Phasitron.[citation needed]

### Volume guarantee

[edit]

Volume guarantee an additional parameter available in many types of ventilators that allows the ventilator to change its inspiratory pressure setting to achieve a minimum tidal volume.[citation needed] This is utilized most often in neonatal patients who need a pressure controlled mode with a consideration for volume control to minimize volutrauma.

## Spontaneous breathing and support settings

[edit]

### Positive end-expiratory pressure

[edit]

Positive end expiratory pressure (PEEP) is pressure applied upon expiration. PEEP is applied using either a valve that is connected to the expiratory port and set manually or a valve managed internally by a mechanical ventilator.

PEEP is a pressure that an exhalation has to bypass, in effect causing alveoli to remain open and not fully deflate. This mechanism for maintaining inflated alveoli helps increase partial pressure of oxygen in arterial blood, and an increase in PEEP increases the PaO2.[17][needs update?]

### Pressure support

[edit]

Pressure support is a spontaneous mode of ventilation also named Pressure Support Ventilation (PSV). The patient initiates every breath and the ventilator delivers support with the preset pressure value. With support from the ventilator, the patient also regulates their own respiratory rate and their tidal volume.

In Pressure Support, the set inspiratory pressure support level is kept constant and there is a decelerating flow. The patient triggers all breaths. If there is a change in the mechanical properties of the lung/thorax and patient effort, the delivered tidal volume will be affected. The user must then regulate the pressure support level to obtain desired ventilation.[18][19]

Pressure support improves oxygenation,[20][needs update?] ventilation and decreases work of breathing.

Also see adaptive support ventilation.

## Other ventilation modes and strategies

[edit]

### Flow-controlled ventilation

[edit]

Flow-controlled ventilation (FCV) is an entirely dynamic ventilation mode, without pauses, with continuous and stable gas flows during both inspiration and expiration, aiming for linear changes in both volume and pressure.[21] FCV is an invasive ventilation mode but, unlike Volume- and pressure controlled modes, it does not rely on a passive expiration created by collapse of the thoracic wall and elastic recoil of the lungs. A high resistant breathing circuit inhibits a passive expiration and therewith allows to fully control and stabilize the expiration flow. FCV creates an inspiration by generating a stable flow from a set End-expiratory pressure (EEP) to a set Peak pressure. Then a stable expiratory flow is created by suctioning.[22][needs update?] This expiratory flow rate is preferably similar to the inspiratory flow, aiming for an I:E ratio of 1:1.0, to minimize energy dissipation in the lungs.[23][medical citation needed][24] FCV is a more efficient ventilation as compared to conventional modes,[25][26][non-primary source needed] allows ventilation through even small lumens (~2 – 10 mm ID)[27][28][non-primary source needed] and results in less applied mechanical power.[25] FCV was invented by Professor Dr. med. Dietmar Enk.[21][non-primary source needed]

### Negative pressure ventilation

[edit]

Main article: Negative pressure ventilator

Negative-pressure ventilation stimulates (or forces) breathing by periodic application of partial vacuum (air pressure reduced below ambient pressure), applied externally to the patient's torso—specifically, chest and abdomen—to assist (or force) the chest to expand, expanding the lungs, resulting in voluntary (or involuntary) inhalation through the patient's airway.[29][30][31]

Various "negative pressure ventilators" (NPVs) have been developed to serve this function—most famously the "Iron lung," a tank in which the patient lays, with only their head exposed to ambient air, while air pressure on the remainder of their body, inside the tank, is varied by pumping, to stimulate chest and lung expansion and contraction. Though not in wide use today, NPVs were the principal forms of hospital and long-term mechanical ventilation in the first half of the 20th century, and remain in limited use today.[29][30][31]

### Closed loop systems

[edit]

#### Adaptive Support Ventilation

[edit]

Adaptive Support Ventilation (ASV) is the only commercially available mode that uses optimal targeting. This technology was patented in 1991[32][33][34] In this positive pressure mode of ventilation, the frequency and tidal volume of breaths of a patient on the ventilator are automatically adjusted and optimized to mimic natural breathing, stimulate spontaneous breathing, and reduce weaning time. In the ASV mode, every breath is synchronized with patient effort if such an effort exists, and otherwise, full mechanical ventilation is provided to the patient.[35][36]

#### Automatic Tube Compensation

[edit]

Automatic Tube Compensation (ATC) is the simplest example of a computer-controlled targeting system on a ventilator. It is a form of servo targeting.[citation needed]

The goal of ATC is to support the resistive work of breathing through the artificial airway.

#### Neurally Adjusted Ventilatory Assist

[edit]

Neurally Adjusted Ventilatory Assist (NAVA) is adjusted by a computer (servo) and is similar to ATC but with more complex requirements for implementation.[citation needed]

In terms of patient-ventilator synchrony, NAVA supports both resistive and elastic work of breathing in proportion to the patient's inspiratory effort

#### Proportional Assist Ventilation

[edit]

Proportional assist ventilation (PAV) is another servo targeting based mode in which the ventilator guarantees the percentage of work regardless of changes in pulmonary compliance and resistance.[37][needs update?]

The ventilator varies the tidal volume and pressure based on the patient's work of breathing. The amount it delivers is proportional to the percentage of assistance it is set to give.

PAV, like NAVA, supports both restrictive and elastic work of breathing in proportion to the patient's inspiratory effort.

### Liquid ventilation

[edit]

Liquid ventilation is a technique of mechanical ventilation in which the lungs are insufflated with an oxygenated perfluorochemical liquid rather than an oxygen-containing gas mixture. The use of perfluorochemicals, rather than nitrogen, as the inert carrier of oxygen and carbon dioxide offers a number of theoretical advantages for the treatment of acute lung injury, including:

* Reducing surface tension by maintaining a fluid interface with alveoli
* Opening of collapsed alveoli by hydraulic pressure with a lower risk of barotrauma
* Providing a reservoir in which oxygen and carbon dioxide can be exchanged with pulmonary capillary blood
* Functioning as a high-efficiency heat exchanger

Despite its theoretical advantages, efficacy studies have been disappointing and the optimal clinical use of LV has yet to be defined.[38][needs update?]

#### Total liquid ventilation

[edit]

In total liquid ventilation (TLV), the entire lung is filled with an oxygenated PFC liquid, and a liquid tidal volume of PFC is actively pumped into and out of the lungs. A specialized apparatus is required to deliver and remove the relatively dense, viscous PFC tidal volumes, and to extracorporeally oxygenate and remove carbon dioxide from the liquid.[39][40][41][needs update?]

#### Partial liquid ventilation

[edit]

In partial liquid ventilation (PLV), the lungs are slowly filled with a volume of PFC equivalent or close to the FRC during gas ventilation. The PFC within the lungs is oxygenated and carbon dioxide is removed by means of gas breaths cycling in the lungs by a conventional gas ventilator.[42][needs update?]

## See also

[edit]

* Medicine portal

* Table of modes of mechanical ventilation
* Mechanical ventilation – Method to mechanically assist or replace spontaneous breathing
* Prone ventilation
* Respiratory therapist – Practitioner in cardio-pulmonary medicine
* Bubble CPAP – Noninvasive form of ventilation for newborns

## References

[edit]

1. ^ Esteban A, Anzueto A, Alía I, Gordo F, Apezteguía C, Pálizas F, et al. (May 2000). "How is mechanical ventilation employed in the intensive care unit? An international utilization review". American Journal of Respiratory and Critical Care Medicine. 161 (5): 1450–1458\. doi:10.1164/ajrccm.161.5.9902018. PMID 10806138.
2. ^ Donn SM (May 2009). "Neonatal ventilators: how do they differ?". Journal of Perinatology. 29 (Suppl 2): S73 – S78. doi:10.1038/jp.2009.23. PMID 19399015.
3. ^ Chatburn RL, Volsko TA, Hazy J, Harris LN, Sanders S (April 2012). "Determining the basis for a taxonomy of mechanical ventilation". Respiratory Care. 57 (4): 514–524\. doi:10.4187/respcare.01327. PMID 22004898. S2CID 27417478.
4. ^ Chatburn RL, El-Khatib M, Mireles-Cabodevila E (November 2014). "A taxonomy for mechanical ventilation: 10 fundamental maxims". Respiratory Care. 59 (11): 1747–1763\. doi:10.4187/respcare.03057. PMID 25118309.
5. ^ Stock MC, Downs JB, Frolicher DA (April 1986). "Airway Pressure Release Ventilation (APRV): A New Ventilatory Support Mode During Acute Lung Injury (ALI)". Critical Care Medicine. 14 (4): 366. doi:10.1097/00003246-198604000-00111. ISSN 1530-0293.
6. ^ "» History of APRV". www.tcavnetwork.org. Archived from the original on 2025-01-17. Retrieved 2025-03-03.
7. ^ Henzler D (2011). "What on earth is APRV?". Critical Care. 15 (1). London, England: 115. doi:10.1186/cc9419. PMC 3222047. PMID 21345265.
8. ^ Andrews P, Shiber J, Madden M, Nieman GF, Camporota L, Habashi NM (2022-07-25). "Myths and Misconceptions of Airway Pressure Release Ventilation: Getting Past the Noise and on to the Signal". Frontiers in Physiology. 13: 928562. doi:10.3389/fphys.2022.928562. PMC 9358044. PMID 35957991.
9. ^ Baum M, Benzer H, Putensen C, Koller W, Putz G (September 1989). "[Biphasic positive airway pressure (BIPAP)--a new form of augmented ventilation]". Der Anaesthesist. 38 (9): 452–458\. PMID 2686487.
10. ^ Putensen C, Zech S, Wrigge H, Zinserling J, Stüber F, Von Spiegel T, et al. (July 2001). "Long-term effects of spontaneous breathing during ventilatory support in patients with acute lung injury". American Journal of Respiratory and Critical Care Medicine. 164 (1): 43–49\. doi:10.1164/ajrccm.164.1.2001078. PMID 11435237.
11. ^ Guthrie SO, Lynn C, Lafleur BJ, Donn SM, Walsh WF (October 2005). "A crossover analysis of mandatory minute ventilation compared to synchronized intermittent mandatory ventilation in neonates". Journal of Perinatology. 25 (10): 643–646\. doi:10.1038/sj.jp.7211371. PMID 16079905.
12. ^ Benzer H (1988) Ventilatory support by intermittent changes in PEEP levels. 4th European Congress on Intensive Care Medicine. Baveno-Stresa
13. ^ Hörmann C, Baum M, Putensen C, Mutz NJ, Benzer H (January 1994). "Biphasic positive airway pressure (BIPAP)--a new mode of ventilatory support". European Journal of Anaesthesiology. 11 (1): 37–42\. PMID 8143712.
14. ^ Levitt MA (November 2001). "A prospective, randomized trial of BiPAP in severe acute congestive heart failure". The Journal of Emergency Medicine. 21 (4): 363–369\. doi:10.1016/s0736-4679(01)00385-7. PMID 11728761.
15. ^ Osadnik CR, Tee VS, Carson-Chahhoud KV, Picot J, Wedzicha JA, Smith BJ (July 2017). "Non-invasive ventilation for the management of acute hypercapnic respiratory failure due to exacerbation of chronic obstructive pulmonary disease". The Cochrane Database of Systematic Reviews. 2017 (7): CD004104. doi:10.1002/14651858.CD004104.pub4. hdl:10044/1/53458. PMC 6483555. PMID 28702957.
16. ^ Allardet-Servent J (December 2011). "High-frequency oscillatory ventilation in adult patients with acute respiratory distress syndrome: where do we stand and where should we go?". Critical Care Medicine. 39 (12): 2761–2762\. doi:10.1097/CCM.0b013e31822a5c35. PMID 22094505.
17. ^ Schuster DP, Klain M, Snyder JV (October 1982). "Comparison of high frequency jet ventilation to conventional ventilation during severe acute respiratory failure in humans". Critical Care Medicine. 10 (10): 625–630\. doi:10.1097/00003246-198210000-00001. PMID 6749433.
18. ^ "Servo-i Mechanical Ventilator". Getinge AB. Retrieved 17 May 2025.
19. ^ "Servo-s Mechanical Ventilator". Getinge AB. Retrieved 17 May 2025.
20. ^ Spieth PM, Carvalho AR, Güldner A, Kasper M, Schubert R, Carvalho NC, et al. (April 2011). "Pressure support improves oxygenation and lung protection compared to pressure-controlled ventilation and is further improved by random variation of pressure support". Critical Care Medicine. 39 (4): 746–755\. doi:10.1097/CCM.0b013e318206bda6. PMID 21263322. S2CID 35876431.
21. ^ a b DE 102016109528A1, Enk D, "Verfahren und Vorrichtung zur Beatmung eines Patienten (method and device for ventilating a patient)", published 7 September 2017, assigned to Current Assignee Ventinova Technologies BV
22. ^ DE 102007013385A1, Enk D, "Gasstromumkehrelement (gas flow reversing element).", published 18 September
23. ^ Barnes T, van Asseldonk D, Enk D (December 2018). "Minimisation of dissipated energy in the airways during mechanical ventilation by using constant inspiratory and expiratory flows - Flow-controlled ventilation (FCV)". Medical Hypotheses. 121: 167–176\. doi:10.1016/j.mehy.2018.09.038. PMID 30396474.
24. ^ Barnes T, Enk D (February 2019). "Ventilation for low dissipated energy achieved using flow control during both inspiration and expiration". Trends in Anaesthesia and Critical Care. 24: 5–12\. doi:10.1016/j.tacc.2018.09.003.
25. ^ a b Weber J, Schmidt J, Straka L, Wirth S, Schumann S (April 2020). "Flow-controlled ventilation improves gas exchange in lung-healthy patients- a randomized interventional cross-over study". Acta Anaesthesiologica Scandinavica. 64 (4): 481–488\. doi:10.1111/aas.13526. PMID 31828755.
26. ^ Sebrechts T, Morrison SG, Schepens T, Saldien V (February 2021). "Flow-controlled ventilation with the Evone ventilator and Tritube versus volume-controlled ventilation: A clinical cross-over pilot study describing oxygenation, ventilation and haemodynamic variables". European Journal of Anaesthesiology. 38 (2): 209–211\. doi:10.1097/EJA.0000000000001326. PMID 33394805.
27. ^ Schmidt J, Günther F, Weber J, Kehm V, Pfeiffer J, Becker C, et al. (December 2019). "Glottic visibility for laryngeal surgery: Tritube vs. microlaryngeal tube: A randomised controlled trial". European Journal of Anaesthesiology. 36 (12): 963–971\. doi:10.1097/EJA.0000000000001110. PMC 6855316. PMID 31644514.
28. ^ Meulemans J, Jans A, Vermeulen K, Vandommele J, Delaere P, Vander Poorten V (2020). "Evone® Flow-Controlled Ventilation During Upper Airway Surgery: A Clinical Feasibility Study and Safety Assessment". Frontiers in Surgery. 7: 6. doi:10.3389/fsurg.2020.00006. PMC 7058692. PMID 32185179.
29. ^ a b Matioc AA (February 2018). "An Anesthesiologist's Perspective on the History of Basic Airway Management: The "Progressive" Era, 1904 to 1960". Anesthesiology. 128 (2): 254–271\. doi:10.1097/ALN.0000000000001975. PMID 29112511. "Early Positive and Alternate Pressure Machines"
30. ^ a b Rockoff, Mark, M.D., "The Iron Lung and Polio,", video (8 minutes), January 11, 2016, OPENPediatrics and Boston Children's Hospital on YouTube, retrieved April 11, 2020 (historical background and images, explanatory diagrams, and live demonstrations)
31. ^ a b Walkey A, Summer R (2008). "E. Noninvasive Mechanical Ventilation". Boston Medical Center ICU Manual (PDF). Boston University. p. 17\. "Negative pressure"
32. ^ US 4,986,268, Tehrani FT, "Method and apparatus for controlling an artificial respiratory.", issued 22 January 1991, assigned to Individual
33. ^ Tehrani FT (October 1991). "Automatic Control of an Artificial Respirator". Proceedings of the Annual International Conference of the IEEE Engineering in Medicine and Biology Society. Vol. 13\. pp. 1738–1739\. doi:10.1109/IEMBS.1991.684729. ISBN 978-0-7803-0216-7. S2CID 63221714.
34. ^ Chatburn RL, Mireles-Cabodevila E (January 2011). "Closed-loop control of mechanical ventilation: description and classification of targeting schemes". Respiratory Care. 56 (1): 85–102\. doi:10.4187/respcare.00967. PMID 21235841.
35. ^ Tehrani FT (December 2008). "Automatic control of mechanical ventilation. Part 1: theory and history of the technology". Journal of Clinical Monitoring and Computing. 22 (6): 409–415\. doi:10.1007/s10877-008-9150-z. PMID 19011976.
36. ^ Tehrani FT (December 2008). "Automatic control of mechanical ventilation. Part 2: the existing techniques and future trends". Journal of Clinical Monitoring and Computing. 22 (6): 417–424\. doi:10.1007/s10877-008-9151-y. PMID 19020981.
37. ^ Younes M (January 1992). "Proportional assist ventilation, a new approach to ventilatory support. Theory". The American Review of Respiratory Disease. 145 (1): 114–120\. doi:10.1164/ajrccm/145.1.114. PMID 1731573.
38. ^ Degraeuwe PL, Vos GD, Blanco CE (October 1995). "Perfluorochemical liquid ventilation: from the animal laboratory to the intensive care unit". The International Journal of Artificial Organs. 18 (10): 674–683\. doi:10.1177/039139889501801020. PMID 8647601. S2CID 13038566.
39. ^ Norris MK, Fuhrman BP, Leach CL (August 1994). "Liquid ventilation: it's not science fiction anymore". AACN Clinical Issues in Critical Care Nursing. 5 (3): 246–254\. doi:10.4037/15597768-1994-3004. PMID 7780839.
40. ^ Greenspan JS (1996). "Physiology and clinical role of liquid ventilation therapy". Journal of Perinatology. 16 (2 Pt 2 Su): S47 – S52. PMID 8732549.
41. ^ Dirkes S (June 1996). "Liquid ventilation: new frontiers in the treatment of ARDS". Critical Care Nurse. 16 (3): 53–58\. doi:10.4037/ccn1996.16.3.53. PMID 8852261.
42. ^ Cox CA, Wolfson MR, Shaffer TH (April 1996). "Liquid ventilation: a comprehensive overview". Neonatal Network. 15 (3): 31–43\. PMID 8715647.

* v
* t
* e

Mechanical ventilation

Fundamentals|

* Modes of mechanical ventilation
* Nomenclature of mechanical ventilation

Modes|

* IMV/SIMV
* CMV
* ACV
* CSV
* PAP
* BPAP/NIV
* CPAP
* PSV
* APRV
* MMV
* PAV
* ASV
* HFV
* List of modes by category

Related illness|

* ARDS
* Atelectotrauma
* Biotrauma
* Pulmonary barotrauma
* Pulmonary volutrauma
* Rheotrauma
* Ventilator-associated pneumonia
* Oxygen toxicity
* Ventilator-associated lung injury

Pressure|

* PEEP
* ΔP
* PIP
* PAW
* Pplat

Volumes|

* VT
* VE
* Vf

Other|

* FiO2
* Cdyn
* Cstatic
* PAO2
* VD/VT
* OI
* A-a gradient
* Mechanical power

*[v]: View this template
*[t]: Discuss this template
*[e]: Edit this template